-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Jan 11 06:48:19 2022 UTC
-- File 20210915_024055_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.432112 to 22.407430
-each separately from 6s to 22s and from 407141us to 432401us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.974740 in 1 packets: 18
MIU packet time from 2021-09-15T02:41:04.432.112.000 to 2021-09-15T02:41:20.407.430.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.974740 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.9551984 to 23.9551457
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999473 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210915_024055_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.3873115 to 8.3876005
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.9746870  -- OK for SL --
   MIU time from 2021-09-15T02:41:04.432.112.000 to 2021-09-15T02:41:04.432.401.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.974740

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.3622875 to 24.3625765
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-15T02:41:20.407.141.000 to 2021-09-15T02:41:20.407.430.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.387311500 corrected by 40bit 10MHz counter of 7.955198400 obtained at 1Hz counter of 6. Result: 6.432114483
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996800
   Min 1st sample MIU time among all SL packets: 2021-09-15T02:41:04.431.971.725
   Max 1st sample MIU time among all SL packets: 2021-09-15T02:41:20.407.289.725
   Min 1st sample XO time among all SL packets: 6.431974207
   Max 1st sample XO time among all SL packets: 22.407291869
   TM values from -255 to 199 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-15T02:41:04.431.971.725 (XO: 6.431974207 - 6.432280208, t=37.2 degC)  Rank: 00083
   TM values: Max =   199  Min =  -255  Ave =    -4.47  Std =    25.40
   Min TM value:-255 at an offset of 98.313 usec -> 2021-09-15T02:41:04.432.070.037 (XO: 6.432072520)
   Max TM value:199 at an offset of 113.700 usec -> 2021-09-15T02:41:04.432.085.425 (XO: 6.432087908)
   Max dTM/dt [1/us]:1.73e+03 at an offset of 70.125 usec -> 2021-09-15T02:41:04.432.041.850 (XO: 6.432044333)
   Min dTM/dt [1/us]:-2.02e+03 at an offset of 98.188 usec -> 2021-09-15T02:41:04.432.069.912 (XO: 6.432072395)

 - Block number: 2 starting at 2021-09-15T02:41:20.407.000.725 (XO: 22.407002868 - 22.407308869, t=37.2 degC)  Rank: 00080
   TM values: Max =   198  Min =  -235  Ave =    -4.51  Std =    20.45
   Min TM value:-235 at an offset of 74.663 usec -> 2021-09-15T02:41:20.407.075.387 (XO: 22.407077531)
   Max TM value:198 at an offset of 90.050 usec -> 2021-09-15T02:41:20.407.090.775 (XO: 22.407092918)
   Max dTM/dt [1/us]:1.70e+03 at an offset of 89.900 usec -> 2021-09-15T02:41:20.407.090.625 (XO: 22.407092768)
   Min dTM/dt [1/us]:-2.05e+03 at an offset of 87.188 usec -> 2021-09-15T02:41:20.407.087.912 (XO: 22.407090056)
