-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jan 10 12:09:06 2022 UTC
-- File 20210913_160905_1227.dat, 18  packets found.
APID:
      1227 (SL) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.963702 to 3.963991
-each separately from 3s to 3s and from 963702us to 963991us.
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU packet time from 2021-09-13T16:09:11.963.702.000 to 2021-09-13T16:09:11.963.991.000
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.8269446 to 4.8269446
   Differences in neigboring packets: 
      0.0000000 in 17 packets (all packets)
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 18 packets (all packets)
Number of generated alerts: 
      0 in 18 packets (all packets)
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20210913_160905_1227.dat, 1 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.7906470 to 5.7909360
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-13T16:09:11.963.702.000 to 2021-09-13T16:09:11.963.991.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.790647000 corrected by 40bit 10MHz counter of 4.826944600 obtained at 1Hz counter of 3. Result: 3.963705580
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-13T16:09:11.963.561.725
   Max 1st sample MIU time among all SL packets: 2021-09-13T16:09:11.963.850.725
   Min 1st sample XO time among all SL packets: 3.963565305
   Max 1st sample XO time among all SL packets: 3.963854306
   TM values from -240 to 195 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-13T16:09:11.963.561.725 (XO: 3.963565305 - 3.963871306, t=37.2 degC)  Rank: 00079
   TM values: Max =   195  Min =  -240  Ave =    -4.52  Std =    26.21
   Min TM value:-240 at an offset of 168.751 usec -> 2021-09-13T16:09:11.963.730.475 (XO: 3.963734055)
   Max TM value:195 at an offset of 125.113 usec -> 2021-09-13T16:09:11.963.686.837 (XO: 3.963690418)
   Max dTM/dt [1/us]:1.62e+03 at an offset of 123.025 usec -> 2021-09-13T16:09:11.963.684.750 (XO: 3.963688330)
   Min dTM/dt [1/us]:-1.83e+03 at an offset of 125.263 usec -> 2021-09-13T16:09:11.963.686.987 (XO: 3.963690568)
