-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jan 10 12:01:52 2022 UTC
-- File 20210913_160554_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.991999 to 22.912442
-each separately from 5s to 22s and from 912153us to 992288us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.919865 in 1 packets: 18
MIU packet time from 2021-09-13T16:06:02.991.999.000 to 2021-09-13T16:06:19.912.442.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.919865 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.8275146 to 23.8274582
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999436 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210913_160554_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.8195144 to 7.8198034
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.9198070  -- OK for SL --
   MIU time from 2021-09-13T16:06:02.991.999.000 to 2021-09-13T16:06:02.992.288.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.919865

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.7396104 to 24.7398994
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-13T16:06:19.912.153.000 to 2021-09-13T16:06:19.912.442.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.819514400 corrected by 40bit 10MHz counter of 6.827514600 obtained at 1Hz counter of 5. Result: 5.992003074
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-13T16:06:02.991.858.725
   Max 1st sample MIU time among all SL packets: 2021-09-13T16:06:19.912.301.725
   Min 1st sample XO time among all SL packets: 5.991862798
   Max 1st sample XO time among all SL packets: 22.912304027
   TM values from -262 to 205 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-13T16:06:02.991.858.725 (XO: 5.991862798 - 5.992168799, t=37.2 degC)  Rank: 00080
   TM values: Max =   204  Min =  -261  Ave =    -4.56  Std =    25.45
   Min TM value:-261 at an offset of 165.363 usec -> 2021-09-13T16:06:02.992.024.088 (XO: 5.992028161)
   Max TM value:204 at an offset of 165.076 usec -> 2021-09-13T16:06:02.992.023.800 (XO: 5.992027874)
   Max dTM/dt [1/us]:1.94e+03 at an offset of 109.513 usec -> 2021-09-13T16:06:02.991.968.237 (XO: 5.991972311)
   Min dTM/dt [1/us]:-2.03e+03 at an offset of 106.975 usec -> 2021-09-13T16:06:02.991.965.700 (XO: 5.991969774)

 - Block number: 2 starting at 2021-09-13T16:06:19.912.012.725 (XO: 22.912015026 - 22.912321027, t=37.2 degC)  Rank: 00085
   TM values: Max =   205  Min =  -262  Ave =    -4.43  Std =    28.44
   Min TM value:-262 at an offset of 118.700 usec -> 2021-09-13T16:06:19.912.131.425 (XO: 22.912133726)
   Max TM value:205 at an offset of 118.375 usec -> 2021-09-13T16:06:19.912.131.100 (XO: 22.912133401)
   Max dTM/dt [1/us]:1.98e+03 at an offset of 121.038 usec -> 2021-09-13T16:06:19.912.133.762 (XO: 22.912136064)
   Min dTM/dt [1/us]:-2.33e+03 at an offset of 118.488 usec -> 2021-09-13T16:06:19.912.131.212 (XO: 22.912133514)
