-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jan 10 09:25:09 2022 UTC
-- File 20210913_142155_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.368488 to 41.577590
-each separately from 8s to 41s and from 368488us to 838168us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.469102 in 1 packets: 18
      13.739133 in 1 packets: 36
MIU packet time from 2021-09-13T14:22:06.368.488.000 to 2021-09-13T14:22:39.577.590.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.469102 in 1 packets: 18
      13.739133 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.9204533 to 42.9203436
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.9999369 in 1 packets: 18
      13.9999534 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210913_142155_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.2889418 to 10.2892308
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.4690390  -- OK for SL --
   MIU time from 2021-09-13T14:22:06.368.488.000 to 2021-09-13T14:22:06.368.777.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.469102

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.7582698 to 29.7585588
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.7390870  -- OK for SL --
   MIU time from 2021-09-13T14:22:25.837.879.000 to 2021-09-13T14:22:25.838.168.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.739133

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.4976458 to 43.4979348
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-13T14:22:39.577.301.000 to 2021-09-13T14:22:39.577.590.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 10.288941800 corrected by 40bit 10MHz counter of 9.920453300 obtained at 1Hz counter of 8. Result: 8.368489716
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-13T14:22:06.368.347.725
   Max 1st sample MIU time among all SL packets: 2021-09-13T14:22:39.577.449.725
   Min 1st sample XO time among all SL packets: 8.368349441
   Max 1st sample XO time among all SL packets: 41.577452831
   TM values from -249 to 236 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-13T14:22:06.368.347.725 (XO: 8.368349441 - 8.368655442, t=37.2 degC)  Rank: 00080
   TM values: Max =   211  Min =  -243  Ave =    -4.34  Std =    18.93
   Min TM value:-243 at an offset of 112.125 usec -> 2021-09-13T14:22:06.368.459.850 (XO: 8.368461566)
   Max TM value:211 at an offset of 111.800 usec -> 2021-09-13T14:22:06.368.459.525 (XO: 8.368461241)
   Max dTM/dt [1/us]:2.10e+03 at an offset of 114.475 usec -> 2021-09-13T14:22:06.368.462.200 (XO: 8.368463916)
   Min dTM/dt [1/us]:-2.31e+03 at an offset of 111.913 usec -> 2021-09-13T14:22:06.368.459.637 (XO: 8.368461353)

 - Block number: 2 starting at 2021-09-13T14:22:25.837.738.725 (XO: 27.837742173 - 27.838048174, t=37.2 degC)  Rank: 00087
   TM values: Max =   236  Min =  -249  Ave =    -4.36  Std =    23.71
   Min TM value:-249 at an offset of 110.963 usec -> 2021-09-13T14:22:25.837.849.687 (XO: 27.837853136)
   Max TM value:236 at an offset of 110.650 usec -> 2021-09-13T14:22:25.837.849.375 (XO: 27.837852824)
   Max dTM/dt [1/us]:2.21e+03 at an offset of 108.638 usec -> 2021-09-13T14:22:25.837.847.362 (XO: 27.837850811)
   Min dTM/dt [1/us]:-2.42e+03 at an offset of 110.775 usec -> 2021-09-13T14:22:25.837.849.500 (XO: 27.837852949)

 - Block number: 3 starting at 2021-09-13T14:22:39.577.160.725 (XO: 41.577163830 - 41.577469831, t=37.2 degC)  Rank: 00079
   TM values: Max =   200  Min =  -239  Ave =    -4.37  Std =    19.11
   Min TM value:-239 at an offset of 123.950 usec -> 2021-09-13T14:22:39.577.284.675 (XO: 41.577287780)
   Max TM value:200 at an offset of 123.675 usec -> 2021-09-13T14:22:39.577.284.400 (XO: 41.577287505)
   Max dTM/dt [1/us]:1.77e+03 at an offset of 126.363 usec -> 2021-09-13T14:22:39.577.287.087 (XO: 41.577290193)
   Min dTM/dt [1/us]:-1.85e+03 at an offset of 123.813 usec -> 2021-09-13T14:22:39.577.284.537 (XO: 41.577287643)
