-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jan 10 05:25:49 2022 UTC
-- File 20210913_111404_1227.dat, 126  packets found.
APID:
      1227 (SL) in 126 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 126 packets (all packets)
Total number of packets: 
      18 in 126 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.849461 to 94.806448
-each separately from 14s to 94s and from 264489us to 849750us.
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.908723 in 1 packets: 18
      23.505727 in 1 packets: 36
      1.169426 in 1 packets: 54
      23.275756 in 1 packets: 72
      6.948694 in 1 packets: 90
      19.146638 in 1 packets: 108
MIU packet time from 2021-09-13T11:14:21.849.461.000 to 2021-09-13T11:15:41.806.448.000
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.908723 in 1 packets: 18
      23.505727 in 1 packets: 36
      1.169426 in 1 packets: 54
      23.275756 in 1 packets: 72
      6.948694 in 1 packets: 90
      19.146638 in 1 packets: 108
MIU orbit number: 
      60000 in 126 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 126 packets (all packets)
MIU orbit information / number of events: 
      0 in 126 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 126 packets (all packets)
Instrument anomalies: 
      0 (OK) in 126 packets (all packets)
FPGA configuration version: 
      211 in 126 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.4320985 to 95.4318328
   Differences in neigboring packets: 
      0.0000000 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.9999801 in 1 packets: 18
      23.9999203 in 1 packets: 36
      0.9999967 in 1 packets: 54
      22.9999236 in 1 packets: 72
      6.9999768 in 1 packets: 90
      18.9999368 in 1 packets: 108
Temperature FPGA: 
      05 -> 48.5 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 72 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 126 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 126 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 126 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 107 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 107 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 126 packets (all packets)
Event memory error: 
      0 in 126 packets (all packets)
Configuration error: 
      0 in 126 packets (all packets)
SDRAM CRC error: 
      0 in 126 packets (all packets)
Radio gain: 
      0 in 126 packets (all packets)
Radio bandwidth: 
      0 in 126 packets (all packets)
Digital gain: 
      0 in 126 packets (all packets)
Analog gain: 
      0 in 126 packets (all packets)
 
SURVEY DATA READER
-- File 20210913_111404_1227.dat, 7 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.2815605 to 16.2818495
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.9087030  -- OK for SL --
   MIU time from 2021-09-13T11:14:21.849.461.000 to 2021-09-13T11:14:21.849.750.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.908723

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.1905525 to 22.1908415
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.5056470  -- OK for SL --
   MIU time from 2021-09-13T11:14:27.758.473.000 to 2021-09-13T11:14:27.758.762.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.505727

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 45.6964885 to 45.6967775
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 1.1694230  -- OK for SL --
   MIU time from 2021-09-13T11:14:51.264.489.000 to 2021-09-13T11:14:51.264.778.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 1.169426

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.8662005 to 46.8664895
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.2756790  -- OK for SL --
   MIU time from 2021-09-13T11:14:52.434.204.000 to 2021-09-13T11:14:52.434.493.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.275756

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 70.1421685 to 70.1424575
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.9486710  -- OK for SL --
   MIU time from 2021-09-13T11:15:15.710.249.000 to 2021-09-13T11:15:15.710.538.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.948694

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 77.0911285 to 77.0914175
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.1465750  -- OK for SL --
   MIU time from 2021-09-13T11:15:22.659.232.000 to 2021-09-13T11:15:22.659.521.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.146638

 - Block number: 7 from the sequence of 18 packets.
   Block numbers in the data:
      7 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 96.2379925 to 96.2382815
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-13T11:15:41.806.159.000 to 2021-09-13T11:15:41.806.448.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 126 points based on 104 XO calibration points at 1Hz counter from 0 to 103
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 16.281560500 corrected by 40bit 10MHz counter of 15.432098500 obtained at 1Hz counter of 14. Result: 14.849464803
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996800
   Min 1st sample MIU time among all SL packets: 2021-09-13T11:14:21.849.320.725
   Max 1st sample MIU time among all SL packets: 2021-09-13T11:15:41.806.307.725
   Min 1st sample XO time among all SL packets: 14.849324528
   Max 1st sample XO time among all SL packets: 94.806311005
   TM values from -235 to 207 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-13T11:14:21.849.320.725 (XO: 14.849324528 - 14.849630529, t=37.2 degC)  Rank: 00077
   TM values: Max =   177  Min =  -224  Ave =    -4.43  Std =    17.64
   Min TM value:-224 at an offset of 126.050 usec -> 2021-09-13T11:14:21.849.446.775 (XO: 14.849450578)
   Max TM value:177 at an offset of 128.588 usec -> 2021-09-13T11:14:21.849.449.312 (XO: 14.849453116)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 128.400 usec -> 2021-09-13T11:14:21.849.449.125 (XO: 14.849452928)
   Min dTM/dt [1/us]:-1.88e+03 at an offset of 125.825 usec -> 2021-09-13T11:14:21.849.446.550 (XO: 14.849450353)

 - Block number: 2 starting at 2021-09-13T11:14:27.758.332.725 (XO: 20.758336203 - 20.758642204, t=37.2 degC)  Rank: 00088
   TM values: Max =   207  Min =  -235  Ave =    -4.41  Std =    17.20
   Min TM value:-235 at an offset of 124.613 usec -> 2021-09-13T11:14:27.758.457.337 (XO: 20.758460816)
   Max TM value:207 at an offset of 127.075 usec -> 2021-09-13T11:14:27.758.459.800 (XO: 20.758463279)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 122.325 usec -> 2021-09-13T11:14:27.758.455.050 (XO: 20.758458529)
   Min dTM/dt [1/us]:-1.95e+03 at an offset of 124.438 usec -> 2021-09-13T11:14:27.758.457.162 (XO: 20.758460641)

 - Block number: 3 starting at 2021-09-13T11:14:51.264.348.725 (XO: 44.264350197 - 44.264656198, t=37.2 degC)  Rank: 00079
   TM values: Max =   169  Min =  -231  Ave =    -4.46  Std =    18.12
   Min TM value:-231 at an offset of 110.325 usec -> 2021-09-13T11:14:51.264.459.050 (XO: 44.264460523)
   Max TM value:169 at an offset of 112.975 usec -> 2021-09-13T11:14:51.264.461.700 (XO: 44.264463173)
   Max dTM/dt [1/us]:1.45e+03 at an offset of 112.763 usec -> 2021-09-13T11:14:51.264.461.487 (XO: 44.264462960)
   Min dTM/dt [1/us]:-1.61e+03 at an offset of 110.113 usec -> 2021-09-13T11:14:51.264.458.837 (XO: 44.264460310)

 - Block number: 4 starting at 2021-09-13T11:14:52.434.063.725 (XO: 45.434066057 - 45.434372058, t=37.2 degC)  Rank: 00076
   TM values: Max =   183  Min =  -234  Ave =    -4.47  Std =    18.94
   Min TM value:-234 at an offset of 127.800 usec -> 2021-09-13T11:14:52.434.191.525 (XO: 45.434193858)
   Max TM value:183 at an offset of 127.475 usec -> 2021-09-13T11:14:52.434.191.200 (XO: 45.434193533)
   Max dTM/dt [1/us]:1.41e+03 at an offset of 125.350 usec -> 2021-09-13T11:14:52.434.189.075 (XO: 45.434191408)
   Min dTM/dt [1/us]:-1.70e+03 at an offset of 127.600 usec -> 2021-09-13T11:14:52.434.191.325 (XO: 45.434193658)

 - Block number: 5 starting at 2021-09-13T11:15:15.710.108.725 (XO: 68.710111368 - 68.710417369, t=37.2 degC)  Rank: 00079
   TM values: Max =   166  Min =  -215  Ave =    -4.47  Std =    17.84
   Min TM value:-215 at an offset of 122.538 usec -> 2021-09-13T11:15:15.710.231.262 (XO: 68.710233906)
   Max TM value:166 at an offset of 125.250 usec -> 2021-09-13T11:15:15.710.233.975 (XO: 68.710236619)
   Max dTM/dt [1/us]:1.32e+03 at an offset of 125.088 usec -> 2021-09-13T11:15:15.710.233.812 (XO: 68.710236456)
   Min dTM/dt [1/us]:-1.60e+03 at an offset of 122.375 usec -> 2021-09-13T11:15:15.710.231.100 (XO: 68.710233744)

 - Block number: 6 starting at 2021-09-13T11:15:22.659.091.725 (XO: 75.659094400 - 75.659400401, t=37.2 degC)  Rank: 00080
   TM values: Max =   172  Min =  -225  Ave =    -4.44  Std =    16.70
   Min TM value:-225 at an offset of 113.625 usec -> 2021-09-13T11:15:22.659.205.350 (XO: 75.659208025)
   Max TM value:172 at an offset of 113.338 usec -> 2021-09-13T11:15:22.659.205.062 (XO: 75.659207738)
   Max dTM/dt [1/us]:1.79e+03 at an offset of 116.038 usec -> 2021-09-13T11:15:22.659.207.762 (XO: 75.659210438)
   Min dTM/dt [1/us]:-1.85e+03 at an offset of 113.438 usec -> 2021-09-13T11:15:22.659.205.162 (XO: 75.659207838)

 - Block number: 7 starting at 2021-09-13T11:15:41.806.018.725 (XO: 94.806022004 - 94.806328005, t=37.2 degC)  Rank: 00076
   TM values: Max =   165  Min =  -224  Ave =    -4.40  Std =    24.01
   Min TM value:-224 at an offset of 109.600 usec -> 2021-09-13T11:15:41.806.128.325 (XO: 94.806131605)
   Max TM value:165 at an offset of 120.713 usec -> 2021-09-13T11:15:41.806.139.437 (XO: 94.806142717)
   Max dTM/dt [1/us]:1.52e+03 at an offset of 125.350 usec -> 2021-09-13T11:15:41.806.144.075 (XO: 94.806147355)
   Min dTM/dt [1/us]:-1.53e+03 at an offset of 122.663 usec -> 2021-09-13T11:15:41.806.141.387 (XO: 94.806144667)
