-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jan 10 03:34:55 2022 UTC
-- File 20210913_095318_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 13.198070 to 23.857464
-each separately from 13s to 23s and from 198070us to 857464us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.658816 in 1 packets: 18
MIU packet time from 2021-09-13T09:53:34.198.070.000 to 2021-09-13T09:53:44.857.464.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.658816 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.9524341 to 24.9524011
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999670 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210913_095318_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.1505050 to 15.1507940
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.6587830  -- OK for SL --
   MIU time from 2021-09-13T09:53:34.198.070.000 to 2021-09-13T09:53:34.198.359.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.658816

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.8095770 to 25.8098660
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-13T09:53:44.857.175.000 to 2021-09-13T09:53:44.857.464.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 13, 32bit 10MHz counter = 15.150505000 corrected by 40bit 10MHz counter of 14.952434100 obtained at 1Hz counter of 13. Result: 13.198071554
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-13T09:53:34.197.929.725
   Max 1st sample MIU time among all SL packets: 2021-09-13T09:53:44.857.323.725
   Min 1st sample XO time among all SL packets: 13.197931278
   Max 1st sample XO time among all SL packets: 23.857327540
   TM values from -232 to 193 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-13T09:53:34.197.929.725 (XO: 13.197931278 - 13.198237279, t=37.2 degC)  Rank: 00079
   TM values: Max =   193  Min =  -217  Ave =    -4.50  Std =    19.66
   Min TM value:-217 at an offset of 109.075 usec -> 2021-09-13T09:53:34.198.038.800 (XO: 13.198040354)
   Max TM value:193 at an offset of 108.750 usec -> 2021-09-13T09:53:34.198.038.475 (XO: 13.198040029)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 111.413 usec -> 2021-09-13T09:53:34.198.041.137 (XO: 13.198042691)
   Min dTM/dt [1/us]:-2.05e+03 at an offset of 108.863 usec -> 2021-09-13T09:53:34.198.038.587 (XO: 13.198040141)

 - Block number: 2 starting at 2021-09-13T09:53:44.857.034.725 (XO: 23.857038539 - 23.857344540, t=37.2 degC)  Rank: 00083
   TM values: Max =   163  Min =  -232  Ave =    -4.59  Std =    19.24
   Min TM value:-232 at an offset of 118.075 usec -> 2021-09-13T09:53:44.857.152.800 (XO: 23.857156614)
   Max TM value:163 at an offset of 120.663 usec -> 2021-09-13T09:53:44.857.155.387 (XO: 23.857159202)
   Max dTM/dt [1/us]:1.42e+03 at an offset of 120.488 usec -> 2021-09-13T09:53:44.857.155.212 (XO: 23.857159027)
   Min dTM/dt [1/us]:-1.54e+03 at an offset of 117.850 usec -> 2021-09-13T09:53:44.857.152.575 (XO: 23.857156389)
