-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jan  9 22:32:22 2022 UTC
-- File 20210912_162336_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.367646 to 28.428475
-each separately from 10s to 28s and from 367646us to 428475us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.060251 in 1 packets: 18
MIU packet time from 2021-09-12T16:23:49.367.646.000 to 2021-09-12T16:24:07.428.475.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.060251 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.8514353 to 29.8513754
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.9999401 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210912_162336_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.2190820 to 12.2193710
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.0601910  -- OK for SL --
   MIU time from 2021-09-12T16:23:49.367.646.000 to 2021-09-12T16:23:49.367.935.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.060251

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.2795620 to 30.2798510
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-12T16:24:07.428.186.000 to 2021-09-12T16:24:07.428.475.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.219082000 corrected by 40bit 10MHz counter of 11.851435300 obtained at 1Hz counter of 10. Result: 10.367647913
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-12T16:23:49.367.505.725
   Max 1st sample MIU time among all SL packets: 2021-09-12T16:24:07.428.334.725
   Min 1st sample XO time among all SL packets: 10.367507638
   Max 1st sample XO time among all SL packets: 28.428336739
   TM values from -261 to 229 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-12T16:23:49.367.505.725 (XO: 10.367507638 - 10.367813639, t=37.2 degC)  Rank: 00081
   TM values: Max =   229  Min =  -261  Ave =    -4.55  Std =    27.08
   Min TM value:-261 at an offset of 123.275 usec -> 2021-09-12T16:23:49.367.629.000 (XO: 10.367630913)
   Max TM value:229 at an offset of 122.988 usec -> 2021-09-12T16:23:49.367.628.712 (XO: 10.367630626)
   Max dTM/dt [1/us]:2.16e+03 at an offset of 125.625 usec -> 2021-09-12T16:23:49.367.631.350 (XO: 10.367633263)
   Min dTM/dt [1/us]:-2.36e+03 at an offset of 123.113 usec -> 2021-09-12T16:23:49.367.628.837 (XO: 10.367630751)

 - Block number: 2 starting at 2021-09-12T16:24:07.428.045.725 (XO: 28.428047738 - 28.428353739, t=37.2 degC)  Rank: 00079
   TM values: Max =   229  Min =  -260  Ave =    -4.56  Std =    23.10
   Min TM value:-260 at an offset of 113.150 usec -> 2021-09-12T16:24:07.428.158.875 (XO: 28.428160888)
   Max TM value:229 at an offset of 110.638 usec -> 2021-09-12T16:24:07.428.156.362 (XO: 28.428158375)
   Max dTM/dt [1/us]:2.40e+03 at an offset of 113.263 usec -> 2021-09-12T16:24:07.428.158.987 (XO: 28.428161000)
   Min dTM/dt [1/us]:-2.52e+03 at an offset of 110.738 usec -> 2021-09-12T16:24:07.428.156.462 (XO: 28.428158475)
