-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jan  9 16:55:26 2022 UTC
-- File 20210912_113652_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.247795 to 20.261200
-each separately from 2s to 20s and from 247795us to 261200us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.012827 in 1 packets: 18
MIU packet time from 2021-09-12T11:36:57.247.795.000 to 2021-09-12T11:37:15.261.200.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.012827 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.9253476 to 21.9252879
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.9999403 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210912_113652_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.1731441 to 4.1734331
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.0127670  -- OK for SL --
   MIU time from 2021-09-12T11:36:57.247.795.000 to 2021-09-12T11:36:57.248.084.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.012827

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.1862001 to 22.1864891
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-12T11:37:15.260.911.000 to 2021-09-12T11:37:15.261.200.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.173144100 corrected by 40bit 10MHz counter of 3.925347600 obtained at 1Hz counter of 2. Result: 2.247797318
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-12T11:36:57.247.654.725
   Max 1st sample MIU time among all SL packets: 2021-09-12T11:37:15.261.059.725
   Min 1st sample XO time among all SL packets: 2.247657042
   Max 1st sample XO time among all SL packets: 20.261061787
   TM values from -238 to 203 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-12T11:36:57.247.654.725 (XO: 2.247657042 - 2.247963043, t=37.2 degC)  Rank: 00079
   TM values: Max =   178  Min =  -230  Ave =    -4.47  Std =    19.66
   Min TM value:-230 at an offset of 118.913 usec -> 2021-09-12T11:36:57.247.773.637 (XO: 2.247775955)
   Max TM value:178 at an offset of 118.613 usec -> 2021-09-12T11:36:57.247.773.337 (XO: 2.247775655)
   Max dTM/dt [1/us]:1.79e+03 at an offset of 121.288 usec -> 2021-09-12T11:36:57.247.776.012 (XO: 2.247778330)
   Min dTM/dt [1/us]:-1.95e+03 at an offset of 118.738 usec -> 2021-09-12T11:36:57.247.773.462 (XO: 2.247775780)

 - Block number: 2 starting at 2021-09-12T11:37:15.260.770.725 (XO: 20.260772786 - 20.261078787, t=37.2 degC)  Rank: 00081
   TM values: Max =   203  Min =  -238  Ave =    -4.45  Std =    27.96
   Min TM value:-238 at an offset of 152.288 usec -> 2021-09-12T11:37:15.260.923.013 (XO: 20.260925074)
   Max TM value:203 at an offset of 123.725 usec -> 2021-09-12T11:37:15.260.894.450 (XO: 20.260896511)
   Max dTM/dt [1/us]:1.83e+03 at an offset of 126.363 usec -> 2021-09-12T11:37:15.260.897.087 (XO: 20.260899148)
   Min dTM/dt [1/us]:-2.17e+03 at an offset of 123.838 usec -> 2021-09-12T11:37:15.260.894.562 (XO: 20.260896623)
