-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jan  9 15:38:03 2022 UTC
-- File 20210912_103130_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.074122 to 16.895084
-each separately from 6s to 16s and from 74122us to 895084us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.820384 in 1 packets: 18
MIU packet time from 2021-09-12T10:31:40.074.122.000 to 2021-09-12T10:31:50.895.084.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.820384 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.2849854 to 18.2849523
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999669 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210912_103130_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.3591084 to 8.3593974
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.8203510  -- OK for SL --
   MIU time from 2021-09-12T10:31:40.074.122.000 to 2021-09-12T10:31:40.074.411.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.820384

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.1797484 to 19.1800374
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-12T10:31:50.894.795.000 to 2021-09-12T10:31:50.895.084.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.359108400 corrected by 40bit 10MHz counter of 8.284985400 obtained at 1Hz counter of 6. Result: 6.074123245
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-12T10:31:40.073.981.725
   Max 1st sample MIU time among all SL packets: 2021-09-12T10:31:50.894.943.725
   Min 1st sample XO time among all SL packets: 6.073982969
   Max 1st sample XO time among all SL packets: 16.894947778
   TM values from -219 to 171 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-12T10:31:40.073.981.725 (XO: 6.073982969 - 6.074288970, t=37.2 degC)  Rank: 00079
   TM values: Max =   156  Min =  -207  Ave =    -4.48  Std =    17.16
   Min TM value:-207 at an offset of 114.275 usec -> 2021-09-12T10:31:40.074.096.000 (XO: 6.074097245)
   Max TM value:156 at an offset of 116.950 usec -> 2021-09-12T10:31:40.074.098.675 (XO: 6.074099920)
   Max dTM/dt [1/us]:1.24e+03 at an offset of 116.788 usec -> 2021-09-12T10:31:40.074.098.512 (XO: 6.074099757)
   Min dTM/dt [1/us]:-1.39e+03 at an offset of 101.550 usec -> 2021-09-12T10:31:40.074.083.275 (XO: 6.074084519)

 - Block number: 2 starting at 2021-09-12T10:31:50.894.654.725 (XO: 16.894658777 - 16.894964778, t=37.2 degC)  Rank: 00083
   TM values: Max =   171  Min =  -219  Ave =    -4.35  Std =    18.97
   Min TM value:-219 at an offset of 126.688 usec -> 2021-09-12T10:31:50.894.781.412 (XO: 16.894785465)
   Max TM value:171 at an offset of 129.225 usec -> 2021-09-12T10:31:50.894.783.950 (XO: 16.894788003)
   Max dTM/dt [1/us]:1.70e+03 at an offset of 129.038 usec -> 2021-09-12T10:31:50.894.783.762 (XO: 16.894787815)
   Min dTM/dt [1/us]:-1.81e+03 at an offset of 126.450 usec -> 2021-09-12T10:31:50.894.781.175 (XO: 16.894785228)
