-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jan  9 04:01:23 2022 UTC
-- File 20210911_130532_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.939933 to 15.558288
-each separately from 0s to 15s and from 557999us to 940222us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.617777 in 1 packets: 18
MIU packet time from 2021-09-11T13:05:35.939.933.000 to 2021-09-11T13:05:50.558.288.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.617777 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.9164371 to 16.9163870
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999499 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210911_130532_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      73 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.8563707 to 2.8566597
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.6177270  -- OK for SL --
   MIU time from 2021-09-11T13:05:35.939.933.000 to 2021-09-11T13:05:35.940.222.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.617777

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.4743867 to 17.4746757
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-11T13:05:50.557.999.000 to 2021-09-11T13:05:50.558.288.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.856370700 corrected by 40bit 10MHz counter of 1.916437100 obtained at 1Hz counter of 0. Result: 0.939936796
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-11T13:05:35.939.792.725
   Max 1st sample MIU time among all SL packets: 2021-09-11T13:05:50.558.147.725
   Min 1st sample XO time among all SL packets: 0.939796520
   Max 1st sample XO time among all SL packets: 15.558150267
   TM values from -253 to 232 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-11T13:05:35.939.792.725 (XO: 0.939796520 - 0.940102521, t=37.2 degC)  Rank: 00073
   TM values: Max =   196  Min =  -248  Ave =    -4.52  Std =    22.25
   Min TM value:-248 at an offset of 120.113 usec -> 2021-09-11T13:05:35.939.912.837 (XO: 0.939916633)
   Max TM value:196 at an offset of 122.700 usec -> 2021-09-11T13:05:35.939.915.425 (XO: 0.939919221)
   Max dTM/dt [1/us]:1.76e+03 at an offset of 122.513 usec -> 2021-09-11T13:05:35.939.915.237 (XO: 0.939919033)
   Min dTM/dt [1/us]:-2.21e+03 at an offset of 119.925 usec -> 2021-09-11T13:05:35.939.912.650 (XO: 0.939916446)

 - Block number: 2 starting at 2021-09-11T13:05:50.557.858.725 (XO: 15.557861266 - 15.558167267, t=37.2 degC)  Rank: 00088
   TM values: Max =   232  Min =  -253  Ave =    -4.58  Std =    21.06
   Min TM value:-253 at an offset of 110.488 usec -> 2021-09-11T13:05:50.557.969.212 (XO: 15.557971754)
   Max TM value:232 at an offset of 107.975 usec -> 2021-09-11T13:05:50.557.966.700 (XO: 15.557969241)
   Max dTM/dt [1/us]:2.35e+03 at an offset of 110.600 usec -> 2021-09-11T13:05:50.557.969.325 (XO: 15.557971866)
   Min dTM/dt [1/us]:-2.50e+03 at an offset of 108.063 usec -> 2021-09-11T13:05:50.557.966.787 (XO: 15.557969329)
