-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan  8 22:39:43 2022 UTC
-- File 20210910_225917_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.734398 to 75.807217
-each separately from 7s to 75s and from 44763us to 846055us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.310076 in 1 packets: 18
      15.040177 in 1 packets: 36
      17.760248 in 1 packets: 54
      25.310725 in 1 packets: 72
      1.649859 in 1 packets: 90
MIU packet time from 2021-09-10T22:59:27.734.398.000 to 2021-09-10T23:00:35.807.217.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.310076 in 1 packets: 18
      15.040177 in 1 packets: 36
      17.760248 in 1 packets: 54
      25.310725 in 1 packets: 72
      1.649859 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.7502262 to 76.7500005
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999701 in 1 packets: 18
      14.9999502 in 1 packets: 36
      16.9999435 in 1 packets: 54
      25.9999138 in 1 packets: 72
      0.9999967 in 1 packets: 90
Temperature FPGA: 
      03 -> 51.1 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 72 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
      2 in 35 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90, 91, ...
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
      2 in 35 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90, 91, ...
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20210910_225917_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      73 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.4846249 to 9.4849139
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.3100470  -- OK for SL --
   MIU time from 2021-09-10T22:59:27.734.398.000 to 2021-09-10T22:59:27.734.687.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.310076

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.7949609 to 17.7952499
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0401270  -- OK for SL --
   MIU time from 2021-09-10T22:59:36.044.763.000 to 2021-09-10T22:59:36.045.052.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.040177

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.8353769 to 32.8356659
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.7601910  -- OK for SL --
   MIU time from 2021-09-10T22:59:51.085.229.000 to 2021-09-10T22:59:51.085.518.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.760248

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 50.5958569 to 50.5961459
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.3106390  -- OK for SL --
   MIU time from 2021-09-10T23:00:08.845.766.000 to 2021-09-10T23:00:08.846.055.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.310725

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 75.9067849 to 75.9070739
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 1.6498550  -- OK for SL --
   MIU time from 2021-09-10T23:00:34.156.780.000 to 2021-09-10T23:00:34.157.069.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 1.649859

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 77.5569289 to 77.5572179
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-10T23:00:35.806.928.000 to 2021-09-10T23:00:35.807.217.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.484624900 corrected by 40bit 10MHz counter of 8.750226200 obtained at 1Hz counter of 7. Result: 7.734401124
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-10T22:59:27.734.257.725
   Max 1st sample MIU time among all SL packets: 2021-09-10T23:00:35.807.076.725
   Min 1st sample XO time among all SL packets: 7.734260848
   Max 1st sample XO time among all SL packets: 75.807079788
   TM values from -248 to 240 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-10T22:59:27.734.257.725 (XO: 7.734260848 - 7.734566849, t=37.2 degC)  Rank: 00073
   TM values: Max =   185  Min =  -239  Ave =    -4.39  Std =    18.31
   Min TM value:-239 at an offset of 116.838 usec -> 2021-09-10T22:59:27.734.374.562 (XO: 7.734377686)
   Max TM value:185 at an offset of 116.525 usec -> 2021-09-10T22:59:27.734.374.250 (XO: 7.734377373)
   Max dTM/dt [1/us]:1.67e+03 at an offset of 114.388 usec -> 2021-09-10T22:59:27.734.372.112 (XO: 7.734375236)
   Min dTM/dt [1/us]:-1.81e+03 at an offset of 116.625 usec -> 2021-09-10T22:59:27.734.374.350 (XO: 7.734377473)

 - Block number: 2 starting at 2021-09-10T22:59:36.044.622.725 (XO: 16.044624472 - 16.044930473, t=37.2 degC)  Rank: 00075
   TM values: Max =   198  Min =  -248  Ave =    -4.47  Std =    18.79
   Min TM value:-248 at an offset of 116.800 usec -> 2021-09-10T22:59:36.044.739.525 (XO: 16.044741273)
   Max TM value:198 at an offset of 116.525 usec -> 2021-09-10T22:59:36.044.739.250 (XO: 16.044740998)
   Max dTM/dt [1/us]:2.08e+03 at an offset of 119.175 usec -> 2021-09-10T22:59:36.044.741.900 (XO: 16.044743648)
   Min dTM/dt [1/us]:-2.16e+03 at an offset of 116.625 usec -> 2021-09-10T22:59:36.044.739.350 (XO: 16.044741098)

 - Block number: 3 starting at 2021-09-10T22:59:51.085.088.725 (XO: 31.085090406 - 31.085396407, t=37.2 degC)  Rank: 00086
   TM values: Max =   211  Min =  -248  Ave =    -4.42  Std =    20.56
   Min TM value:-248 at an offset of 128.513 usec -> 2021-09-10T22:59:51.085.217.237 (XO: 31.085218919)
   Max TM value:211 at an offset of 125.975 usec -> 2021-09-10T22:59:51.085.214.700 (XO: 31.085216381)
   Max dTM/dt [1/us]:2.25e+03 at an offset of 128.625 usec -> 2021-09-10T22:59:51.085.217.350 (XO: 31.085219031)
   Min dTM/dt [1/us]:-2.56e+03 at an offset of 126.100 usec -> 2021-09-10T22:59:51.085.214.825 (XO: 31.085216506)

 - Block number: 4 starting at 2021-09-10T23:00:08.845.625.725 (XO: 48.845629416 - 48.845935417, t=37.2 degC)  Rank: 00074
   TM values: Max =   190  Min =  -212  Ave =    -4.40  Std =    17.77
   Min TM value:-212 at an offset of 126.763 usec -> 2021-09-10T23:00:08.845.752.487 (XO: 48.845756178)
   Max TM value:190 at an offset of 129.450 usec -> 2021-09-10T23:00:08.845.755.175 (XO: 48.845758866)
   Max dTM/dt [1/us]:1.68e+03 at an offset of 129.313 usec -> 2021-09-10T23:00:08.845.755.037 (XO: 48.845758729)
   Min dTM/dt [1/us]:-1.42e+03 at an offset of 126.550 usec -> 2021-09-10T23:00:08.845.752.275 (XO: 48.845755966)

 - Block number: 5 starting at 2021-09-10T23:00:34.156.639.725 (XO: 74.156641342 - 74.156947343, t=37.2 degC)  Rank: 00083
   TM values: Max =   240  Min =  -235  Ave =    -4.56  Std =    19.76
   Min TM value:-235 at an offset of 127.138 usec -> 2021-09-10T23:00:34.156.766.862 (XO: 74.156768480)
   Max TM value:240 at an offset of 126.888 usec -> 2021-09-10T23:00:34.156.766.612 (XO: 74.156768230)
   Max dTM/dt [1/us]:2.12e+03 at an offset of 124.875 usec -> 2021-09-10T23:00:34.156.764.600 (XO: 74.156766217)
   Min dTM/dt [1/us]:-2.36e+03 at an offset of 126.988 usec -> 2021-09-10T23:00:34.156.766.712 (XO: 74.156768330)

 - Block number: 6 starting at 2021-09-10T23:00:35.806.787.725 (XO: 75.806790787 - 75.807096788, t=37.2 degC)  Rank: 00080
   TM values: Max =   194  Min =  -230  Ave =    -4.55  Std =    21.17
   Min TM value:-230 at an offset of 101.850 usec -> 2021-09-10T23:00:35.806.889.575 (XO: 75.806892638)
   Max TM value:194 at an offset of 117.325 usec -> 2021-09-10T23:00:35.806.905.050 (XO: 75.806908113)
   Max dTM/dt [1/us]:1.50e+03 at an offset of 117.138 usec -> 2021-09-10T23:00:35.806.904.862 (XO: 75.806907925)
   Min dTM/dt [1/us]:-2.02e+03 at an offset of 114.525 usec -> 2021-09-10T23:00:35.806.902.250 (XO: 75.806905313)
