-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan  8 22:13:43 2022 UTC
-- File 20210910_223108_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.680244 to 81.024503
-each separately from 8s to 81s and from 24214us to 982987us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.060388 in 1 packets: 18
      17.900602 in 1 packets: 36
      16.340597 in 1 packets: 54
      14.380561 in 1 packets: 72
      12.660377 in 1 packets: 90
MIU packet time from 2021-09-10T22:31:19.680.244.000 to 2021-09-10T22:32:32.024.503.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.060388 in 1 packets: 18
      17.900602 in 1 packets: 36
      16.340597 in 1 packets: 54
      14.380561 in 1 packets: 72
      12.660377 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.7419352 to 82.7416925
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.9999635 in 1 packets: 18
      17.9999401 in 1 packets: 36
      15.9999469 in 1 packets: 54
      14.9999501 in 1 packets: 72
      12.9999567 in 1 packets: 90
Temperature FPGA: 
      04 -> 49.8 degC in 90 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      03 -> 51.1 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55
      2 in 53 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55
      2 in 53 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20210910_223108_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.4221800 to 10.4224690
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.0603510  -- OK for SL --
   MIU time from 2021-09-10T22:31:19.680.244.000 to 2021-09-10T22:31:19.680.533.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.060388

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.4828200 to 21.4831090
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.9005430  -- OK for SL --
   MIU time from 2021-09-10T22:31:30.740.921.000 to 2021-09-10T22:31:30.741.210.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.900602

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.3836520 to 39.3839410
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.3405430  -- OK for SL --
   MIU time from 2021-09-10T22:31:48.641.812.000 to 2021-09-10T22:31:48.642.101.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.340597

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 55.7244840 to 55.7247730
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.3805110  -- OK for SL --
   MIU time from 2021-09-10T22:32:04.982.698.000 to 2021-09-10T22:32:04.982.987.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.380561

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 70.1052840 to 70.1055730
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.6603350  -- OK for SL --
   MIU time from 2021-09-10T22:32:19.363.548.000 to 2021-09-10T22:32:19.363.837.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.660377

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 82.7659080 to 82.7661970
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-10T22:32:32.024.214.000 to 2021-09-10T22:32:32.024.503.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 10.422180000 corrected by 40bit 10MHz counter of 9.741935200 obtained at 1Hz counter of 8. Result: 8.680247045
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-10T22:31:19.680.103.725
   Max 1st sample MIU time among all SL packets: 2021-09-10T22:32:32.024.362.725
   Min 1st sample XO time among all SL packets: 8.680106769
   Max 1st sample XO time among all SL packets: 81.024364305
   TM values from -272 to 223 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-10T22:31:19.680.103.725 (XO: 8.680106769 - 8.680412770, t=37.2 degC)  Rank: 00080
   TM values: Max =   217  Min =  -232  Ave =    -4.43  Std =    21.20
   Min TM value:-232 at an offset of 119.625 usec -> 2021-09-10T22:31:19.680.223.350 (XO: 8.680226395)
   Max TM value:217 at an offset of 119.350 usec -> 2021-09-10T22:31:19.680.223.075 (XO: 8.680226120)
   Max dTM/dt [1/us]:1.98e+03 at an offset of 122.000 usec -> 2021-09-10T22:31:19.680.225.725 (XO: 8.680228770)
   Min dTM/dt [1/us]:-2.18e+03 at an offset of 119.450 usec -> 2021-09-10T22:31:19.680.223.175 (XO: 8.680226220)

 - Block number: 2 starting at 2021-09-10T22:31:30.740.780.725 (XO: 19.740783470 - 19.741089471, t=37.2 degC)  Rank: 00076
   TM values: Max =   223  Min =  -272  Ave =    -4.54  Std =    23.13
   Min TM value:-272 at an offset of 122.538 usec -> 2021-09-10T22:31:30.740.903.262 (XO: 19.740906007)
   Max TM value:223 at an offset of 122.263 usec -> 2021-09-10T22:31:30.740.902.987 (XO: 19.740905732)
   Max dTM/dt [1/us]:1.98e+03 at an offset of 124.888 usec -> 2021-09-10T22:31:30.740.905.612 (XO: 19.740908357)
   Min dTM/dt [1/us]:-2.14e+03 at an offset of 122.350 usec -> 2021-09-10T22:31:30.740.903.075 (XO: 19.740905820)

 - Block number: 3 starting at 2021-09-10T22:31:48.641.671.725 (XO: 37.641675043 - 37.641981044, t=37.2 degC)  Rank: 00085
   TM values: Max =   204  Min =  -230  Ave =    -4.51  Std =    19.02
   Min TM value:-230 at an offset of 118.875 usec -> 2021-09-10T22:31:48.641.790.600 (XO: 37.641793918)
   Max TM value:204 at an offset of 118.538 usec -> 2021-09-10T22:31:48.641.790.262 (XO: 37.641793580)
   Max dTM/dt [1/us]:2.11e+03 at an offset of 121.225 usec -> 2021-09-10T22:31:48.641.792.950 (XO: 37.641796268)
   Min dTM/dt [1/us]:-2.44e+03 at an offset of 118.675 usec -> 2021-09-10T22:31:48.641.790.400 (XO: 37.641793718)

 - Block number: 4 starting at 2021-09-10T22:32:04.982.557.725 (XO: 53.982561267 - 53.982867268, t=37.2 degC)  Rank: 00082
   TM values: Max =   216  Min =  -238  Ave =    -4.55  Std =    19.31
   Min TM value:-238 at an offset of 124.375 usec -> 2021-09-10T22:32:04.982.682.100 (XO: 53.982685643)
   Max TM value:216 at an offset of 121.825 usec -> 2021-09-10T22:32:04.982.679.550 (XO: 53.982683093)
   Max dTM/dt [1/us]:2.22e+03 at an offset of 124.463 usec -> 2021-09-10T22:32:04.982.682.187 (XO: 53.982685730)
   Min dTM/dt [1/us]:-2.40e+03 at an offset of 121.913 usec -> 2021-09-10T22:32:04.982.679.637 (XO: 53.982683180)

 - Block number: 5 starting at 2021-09-10T22:32:19.363.407.725 (XO: 68.363409124 - 68.363715125, t=37.2 degC)  Rank: 00076
   TM values: Max =   190  Min =  -223  Ave =    -4.52  Std =    21.57
   Min TM value:-223 at an offset of 107.513 usec -> 2021-09-10T22:32:19.363.515.237 (XO: 68.363516637)
   Max TM value:190 at an offset of 94.675 usec -> 2021-09-10T22:32:19.363.502.400 (XO: 68.363503800)
   Max dTM/dt [1/us]:1.50e+03 at an offset of 110.063 usec -> 2021-09-10T22:32:19.363.517.787 (XO: 68.363519187)
   Min dTM/dt [1/us]:-2.07e+03 at an offset of 94.763 usec -> 2021-09-10T22:32:19.363.502.487 (XO: 68.363503887)

 - Block number: 6 starting at 2021-09-10T22:32:32.024.073.725 (XO: 81.024075304 - 81.024381305, t=37.2 degC)  Rank: 00080
   TM values: Max =   216  Min =  -241  Ave =    -4.63  Std =    18.66
   Min TM value:-241 at an offset of 122.188 usec -> 2021-09-10T22:32:32.024.195.912 (XO: 81.024197492)
   Max TM value:216 at an offset of 121.913 usec -> 2021-09-10T22:32:32.024.195.637 (XO: 81.024197217)
   Max dTM/dt [1/us]:2.30e+03 at an offset of 124.600 usec -> 2021-09-10T22:32:32.024.198.325 (XO: 81.024199905)
   Min dTM/dt [1/us]:-2.18e+03 at an offset of 122.025 usec -> 2021-09-10T22:32:32.024.195.750 (XO: 81.024197330)
