-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan  8 18:25:00 2022 UTC
-- File 20210910_141007_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.973084 to 24.340688
-each separately from 9s to 24s and from 340399us to 973373us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.367026 in 1 packets: 18
MIU packet time from 2021-09-10T14:10:19.973.084.000 to 2021-09-10T14:10:34.340.688.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.367026 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.8601437 to 25.8600937
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999500 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210910_141007_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.8332286 to 11.8335176
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.3669750  -- OK for SL --
   MIU time from 2021-09-10T14:10:19.973.084.000 to 2021-09-10T14:10:19.973.373.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.367026

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.2004926 to 26.2007816
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-10T14:10:34.340.399.000 to 2021-09-10T14:10:34.340.688.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.833228600 corrected by 40bit 10MHz counter of 10.860143700 obtained at 1Hz counter of 9. Result: 9.973088111
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-10T14:10:19.972.943.725
   Max 1st sample MIU time among all SL packets: 2021-09-10T14:10:34.340.547.725
   Min 1st sample XO time among all SL packets: 9.972947836
   Max 1st sample XO time among all SL packets: 24.340548749
   TM values from -239 to 233 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-10T14:10:19.972.943.725 (XO: 9.972947836 - 9.973253837, t=37.2 degC)  Rank: 00083
   TM values: Max =   233  Min =  -239  Ave =    -4.55  Std =    25.74
   Min TM value:-239 at an offset of 130.363 usec -> 2021-09-10T14:10:19.973.074.087 (XO: 9.973078199)
   Max TM value:233 at an offset of 130.125 usec -> 2021-09-10T14:10:19.973.073.850 (XO: 9.973077961)
   Max dTM/dt [1/us]:2.18e+03 at an offset of 132.775 usec -> 2021-09-10T14:10:19.973.076.500 (XO: 9.973080611)
   Min dTM/dt [1/us]:-2.61e+03 at an offset of 130.238 usec -> 2021-09-10T14:10:19.973.073.962 (XO: 9.973078074)

 - Block number: 2 starting at 2021-09-10T14:10:34.340.258.725 (XO: 24.340259748 - 24.340565749, t=37.2 degC)  Rank: 00081
   TM values: Max =   217  Min =  -228  Ave =    -4.54  Std =    16.65
   Min TM value:-228 at an offset of 116.000 usec -> 2021-09-10T14:10:34.340.374.725 (XO: 24.340375748)
   Max TM value:217 at an offset of 115.713 usec -> 2021-09-10T14:10:34.340.374.437 (XO: 24.340375461)
   Max dTM/dt [1/us]:1.80e+03 at an offset of 113.725 usec -> 2021-09-10T14:10:34.340.372.450 (XO: 24.340373473)
   Min dTM/dt [1/us]:-2.26e+03 at an offset of 115.838 usec -> 2021-09-10T14:10:34.340.374.562 (XO: 24.340375586)
