-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan  8 15:39:25 2022 UTC
-- File 20210910_120907_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.005991 to 22.542070
-each separately from 8s to 22s and from 5991us to 542070us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.535501 in 1 packets: 18
MIU packet time from 2021-09-10T12:09:18.005.991.000 to 2021-09-10T12:09:32.542.070.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.535501 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.9054443 to 23.9053979
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999536 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210910_120907_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.9114360 to 9.9117250
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.5354550  -- OK for SL --
   MIU time from 2021-09-10T12:09:18.005.991.000 to 2021-09-10T12:09:18.006.280.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.535501

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.4471800 to 24.4474690
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-10T12:09:32.541.781.000 to 2021-09-10T12:09:32.542.070.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 9.911436000 corrected by 40bit 10MHz counter of 9.905444300 obtained at 1Hz counter of 8. Result: 8.005991720
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-10T12:09:18.005.850.725
   Max 1st sample MIU time among all SL packets: 2021-09-10T12:09:32.541.929.725
   Min 1st sample XO time among all SL packets: 8.005851444
   Max 1st sample XO time among all SL packets: 22.541932613
   TM values from -247 to 247 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-10T12:09:18.005.850.725 (XO: 8.005851444 - 8.006157445, t=37.2 degC)  Rank: 00081
   TM values: Max =   244  Min =  -237  Ave =    -4.55  Std =    17.41
   Min TM value:-237 at an offset of 84.863 usec -> 2021-09-10T12:09:18.005.935.587 (XO: 8.005936307)
   Max TM value:244 at an offset of 84.600 usec -> 2021-09-10T12:09:18.005.935.325 (XO: 8.005936045)
   Max dTM/dt [1/us]:2.12e+03 at an offset of 82.600 usec -> 2021-09-10T12:09:18.005.933.325 (XO: 8.005934045)
   Min dTM/dt [1/us]:-2.34e+03 at an offset of 84.713 usec -> 2021-09-10T12:09:18.005.935.437 (XO: 8.005936157)

 - Block number: 2 starting at 2021-09-10T12:09:32.541.640.725 (XO: 22.541643612 - 22.541949613, t=37.2 degC)  Rank: 00080
   TM values: Max =   247  Min =  -247  Ave =    -4.43  Std =    18.64
   Min TM value:-247 at an offset of 122.500 usec -> 2021-09-10T12:09:32.541.763.225 (XO: 22.541766113)
   Max TM value:247 at an offset of 120.400 usec -> 2021-09-10T12:09:32.541.761.125 (XO: 22.541764013)
   Max dTM/dt [1/us]:1.97e+03 at an offset of 120.275 usec -> 2021-09-10T12:09:32.541.761.000 (XO: 22.541763888)
   Min dTM/dt [1/us]:-2.20e+03 at an offset of 122.363 usec -> 2021-09-10T12:09:32.541.763.087 (XO: 22.541765975)
