-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan  8 14:47:20 2022 UTC
-- File 20210910_113152_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.738747 to 38.708956
-each separately from 9s to 38s and from 708667us to 913253us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.173928 in 1 packets: 18
      11.795414 in 1 packets: 36
MIU packet time from 2021-09-10T11:32:04.738.747.000 to 2021-09-10T11:32:33.708.956.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.173928 in 1 packets: 18
      11.795414 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.8886553 to 39.8885594
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999437 in 1 packets: 18
      11.9999604 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 53 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 53 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210910_113152_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.6274035 to 11.6276925
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.1738710  -- OK for SL --
   MIU time from 2021-09-10T11:32:04.738.747.000 to 2021-09-10T11:32:04.739.036.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.173928

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.8015635 to 28.8018525
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.7953750  -- OK for SL --
   MIU time from 2021-09-10T11:32:21.912.964.000 to 2021-09-10T11:32:21.913.253.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.795414

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 40.5972275 to 40.5975165
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-10T11:32:33.708.667.000 to 2021-09-10T11:32:33.708.956.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.627403500 corrected by 40bit 10MHz counter of 10.888655300 obtained at 1Hz counter of 9. Result: 9.738750638
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996800
   Min 1st sample MIU time among all SL packets: 2021-09-10T11:32:04.738.606.725
   Max 1st sample MIU time among all SL packets: 2021-09-10T11:32:33.708.815.725
   Min 1st sample XO time among all SL packets: 9.738610362
   Max 1st sample XO time among all SL packets: 38.708819164
   TM values from -246 to 227 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-10T11:32:04.738.606.725 (XO: 9.738610362 - 9.738916363, t=37.2 degC)  Rank: 00085
   TM values: Max =   227  Min =  -240  Ave =    -4.41  Std =    19.86
   Min TM value:-240 at an offset of 127.500 usec -> 2021-09-10T11:32:04.738.734.225 (XO: 9.738737863)
   Max TM value:227 at an offset of 127.188 usec -> 2021-09-10T11:32:04.738.733.912 (XO: 9.738737550)
   Max dTM/dt [1/us]:1.89e+03 at an offset of 129.838 usec -> 2021-09-10T11:32:04.738.736.562 (XO: 9.738740200)
   Min dTM/dt [1/us]:-1.97e+03 at an offset of 127.288 usec -> 2021-09-10T11:32:04.738.734.012 (XO: 9.738737650)

 - Block number: 2 starting at 2021-09-10T11:32:21.912.823.725 (XO: 26.912827146 - 26.913133147, t=37.2 degC)  Rank: 00076
   TM values: Max =   197  Min =  -238  Ave =    -4.47  Std =    18.76
   Min TM value:-238 at an offset of 130.263 usec -> 2021-09-10T11:32:21.912.953.987 (XO: 26.912957409)
   Max TM value:197 at an offset of 129.963 usec -> 2021-09-10T11:32:21.912.953.687 (XO: 26.912957109)
   Max dTM/dt [1/us]:1.66e+03 at an offset of 127.838 usec -> 2021-09-10T11:32:21.912.951.562 (XO: 26.912954984)
   Min dTM/dt [1/us]:-1.62e+03 at an offset of 130.063 usec -> 2021-09-10T11:32:21.912.953.787 (XO: 26.912957209)

 - Block number: 3 starting at 2021-09-10T11:32:33.708.526.725 (XO: 38.708530163 - 38.708836164, t=37.2 degC)  Rank: 00078
   TM values: Max =   202  Min =  -246  Ave =    -4.47  Std =    23.79
   Min TM value:-246 at an offset of 115.613 usec -> 2021-09-10T11:32:33.708.642.337 (XO: 38.708645776)
   Max TM value:202 at an offset of 115.300 usec -> 2021-09-10T11:32:33.708.642.025 (XO: 38.708645464)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 117.950 usec -> 2021-09-10T11:32:33.708.644.675 (XO: 38.708648114)
   Min dTM/dt [1/us]:-1.88e+03 at an offset of 115.413 usec -> 2021-09-10T11:32:33.708.642.137 (XO: 38.708645576)
