-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan  8 11:47:27 2022 UTC
-- File 20210910_053014_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.405326 to 85.205893
-each separately from 6s to 85s and from 137343us to 957585us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.551681 in 1 packets: 18
      23.179758 in 1 packets: 36
      15.088512 in 1 packets: 54
      10.634289 in 1 packets: 72
      19.344593 in 1 packets: 90
MIU packet time from 2021-09-10T05:30:23.405.326.000 to 2021-09-10T05:31:42.205.893.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.551681 in 1 packets: 18
      23.179758 in 1 packets: 36
      15.088512 in 1 packets: 54
      10.634289 in 1 packets: 72
      19.344593 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.1626643 to 87.1624023
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999669 in 1 packets: 18
      23.9999205 in 1 packets: 36
      14.9999503 in 1 packets: 54
      9.9999668 in 1 packets: 72
      19.9999335 in 1 packets: 90
Temperature FPGA: 
      04 -> 49.8 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      03 -> 51.1 degC in 36 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Temperature XO: 
      C1 -> 37.2 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55
      2 in 53 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55
      2 in 53 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20210910_053014_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      63 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.5679914 to 8.5682804
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.5516470  -- OK for SL --
   MIU time from 2021-09-10T05:30:23.405.326.000 to 2021-09-10T05:30:23.405.615.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.551681

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      62 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.1199274 to 19.1202164
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.1796790  -- OK for SL --
   MIU time from 2021-09-10T05:30:33.957.296.000 to 2021-09-10T05:30:33.957.585.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.179758

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      120 in 18 packets (all packets)  -- OK --
   10 MHz counter from 42.2998954 to 42.3001844
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0884630  -- OK for SL --
   MIU time from 2021-09-10T05:30:57.137.343.000 to 2021-09-10T05:30:57.137.632.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.088512

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      106 in 18 packets (all packets)  -- OK --
   10 MHz counter from 57.3886474 to 57.3889364
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.6342550  -- OK for SL --
   MIU time from 2021-09-10T05:31:12.226.144.000 to 2021-09-10T05:31:12.226.433.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.634289

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      61 in 18 packets (all packets)  -- OK --
   10 MHz counter from 68.0231914 to 68.0234804
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.3445270  -- OK for SL --
   MIU time from 2021-09-10T05:31:22.860.722.000 to 2021-09-10T05:31:22.861.011.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.344593

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      65 in 18 packets (all packets)  -- OK --
   10 MHz counter from 87.3680074 to 87.3682964
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-10T05:31:42.205.604.000 to 2021-09-10T05:31:42.205.893.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.567991400 corrected by 40bit 10MHz counter of 8.162664300 obtained at 1Hz counter of 6. Result: 6.405328438
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-10T05:30:23.405.185.725
   Max 1st sample MIU time among all SL packets: 2021-09-10T05:31:42.205.752.725
   Min 1st sample XO time among all SL packets: 6.405188162
   Max 1st sample XO time among all SL packets: 85.205754504
   TM values from -210 to 286 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-10T05:30:23.405.185.725 (XO: 6.405188162 - 6.405494163, t=37.2 degC)  Rank: 00063
   TM values: Max =   162  Min =  -194  Ave =    -4.57  Std =    21.06
   Min TM value:-194 at an offset of 127.075 usec -> 2021-09-10T05:30:23.405.312.800 (XO: 6.405315238)
   Max TM value:162 at an offset of 129.588 usec -> 2021-09-10T05:30:23.405.315.312 (XO: 6.405317750)
   Max dTM/dt [1/us]:1.60e+03 at an offset of 129.463 usec -> 2021-09-10T05:30:23.405.315.187 (XO: 6.405317625)
   Min dTM/dt [1/us]:-1.76e+03 at an offset of 126.900 usec -> 2021-09-10T05:30:23.405.312.625 (XO: 6.405315063)

 - Block number: 2 starting at 2021-09-10T05:30:33.957.155.725 (XO: 16.957159084 - 16.957465085, t=37.2 degC)  Rank: 00062
   TM values: Max =   163  Min =  -210  Ave =    -4.44  Std =    20.62
   Min TM value:-210 at an offset of 189.426 usec -> 2021-09-10T05:30:33.957.345.150 (XO: 16.957348509)
   Max TM value:163 at an offset of 191.926 usec -> 2021-09-10T05:30:33.957.347.650 (XO: 16.957351009)
   Max dTM/dt [1/us]:1.51e+03 at an offset of 191.776 usec -> 2021-09-10T05:30:33.957.347.500 (XO: 16.957350859)
   Min dTM/dt [1/us]:-1.54e+03 at an offset of 130.425 usec -> 2021-09-10T05:30:33.957.286.150 (XO: 16.957289509)

 - Block number: 3 starting at 2021-09-10T05:30:57.137.202.725 (XO: 40.137203878 - 40.137509879, t=37.2 degC)  Rank: 00120
   TM values: Max =   175  Min =  -143  Ave =    -4.64  Std =    25.08
   Min TM value:-143 at an offset of 105.825 usec -> 2021-09-10T05:30:57.137.308.550 (XO: 40.137309703)
   Max TM value:175 at an offset of 102.050 usec -> 2021-09-10T05:30:57.137.304.775 (XO: 40.137305928)
   Max dTM/dt [1/us]:3.11e+02 at an offset of 215.401 usec -> 2021-09-10T05:30:57.137.418.125 (XO: 40.137419278)
   Min dTM/dt [1/us]:-3.07e+02 at an offset of 175.013 usec -> 2021-09-10T05:30:57.137.377.738 (XO: 40.137378891)

 - Block number: 4 starting at 2021-09-10T05:31:12.226.003.725 (XO: 55.226005893 - 55.226311894, t=37.2 degC)  Rank: 00106
   TM values: Max =   286  Min =  -147  Ave =    -4.60  Std =    25.26
   Min TM value:-147 at an offset of 122.738 usec -> 2021-09-10T05:31:12.226.126.462 (XO: 55.226128631)
   Max TM value:286 at an offset of 116.950 usec -> 2021-09-10T05:31:12.226.120.675 (XO: 55.226122844)
   Max dTM/dt [1/us]:1.06e+03 at an offset of 116.888 usec -> 2021-09-10T05:31:12.226.120.612 (XO: 55.226122781)
   Min dTM/dt [1/us]:-1.31e+03 at an offset of 117.263 usec -> 2021-09-10T05:31:12.226.120.987 (XO: 55.226123156)

 - Block number: 5 starting at 2021-09-10T05:31:22.860.581.725 (XO: 65.860585251 - 65.860891252, t=37.2 degC)  Rank: 00061
   TM values: Max =   147  Min =  -178  Ave =    -4.59  Std =    16.20
   Min TM value:-178 at an offset of 115.175 usec -> 2021-09-10T05:31:22.860.696.900 (XO: 65.860700426)
   Max TM value:147 at an offset of 117.750 usec -> 2021-09-10T05:31:22.860.699.475 (XO: 65.860703001)
   Max dTM/dt [1/us]:1.10e+03 at an offset of 117.575 usec -> 2021-09-10T05:31:22.860.699.300 (XO: 65.860702826)
   Min dTM/dt [1/us]:-1.25e+03 at an offset of 114.925 usec -> 2021-09-10T05:31:22.860.696.650 (XO: 65.860700176)

 - Block number: 6 starting at 2021-09-10T05:31:42.205.463.725 (XO: 85.205465503 - 85.205771504, t=37.2 degC)  Rank: 00065
   TM values: Max =   161  Min =  -196  Ave =    -4.45  Std =    16.09
   Min TM value:-196 at an offset of 130.038 usec -> 2021-09-10T05:31:42.205.593.762 (XO: 85.205595541)
   Max TM value:161 at an offset of 132.538 usec -> 2021-09-10T05:31:42.205.596.262 (XO: 85.205598041)
   Max dTM/dt [1/us]:1.54e+03 at an offset of 132.425 usec -> 2021-09-10T05:31:42.205.596.150 (XO: 85.205597928)
   Min dTM/dt [1/us]:-1.43e+03 at an offset of 129.863 usec -> 2021-09-10T05:31:42.205.593.587 (XO: 85.205595366)
