-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan  8 11:19:49 2022 UTC
-- File 20210910_044322_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.338171 to 48.208618
-each separately from 10s to 48s and from 163611us to 968917us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.630168 in 1 packets: 18
      11.194694 in 1 packets: 36
      9.044429 in 1 packets: 54
MIU packet time from 2021-09-10T04:43:35.338.171.000 to 2021-09-10T04:44:13.208.618.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.630168 in 1 packets: 18
      11.194694 in 1 packets: 36
      9.044429 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.7971634 to 49.7970370
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999434 in 1 packets: 18
      11.9999601 in 1 packets: 36
      8.9999701 in 1 packets: 54
Temperature FPGA: 
      04 -> 49.8 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210910_044322_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      120 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.1353351 to 12.1356241
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.6301110  -- OK for SL --
   MIU time from 2021-09-10T04:43:35.338.171.000 to 2021-09-10T04:43:35.338.460.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.630168

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      57 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.7657351 to 29.7660241
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.1946550  -- OK for SL --
   MIU time from 2021-09-10T04:43:52.968.628.000 to 2021-09-10T04:43:52.968.917.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.194694

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      108 in 18 packets (all packets)  -- OK --
   10 MHz counter from 40.9606791 to 40.9609681
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.0443990  -- OK for SL --
   MIU time from 2021-09-10T04:44:04.163.611.000 to 2021-09-10T04:44:04.163.900.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.044429

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      54 in 18 packets (all packets)  -- OK --
   10 MHz counter from 50.0053671 to 50.0056561
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-10T04:44:13.208.329.000 to 2021-09-10T04:44:13.208.618.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.135335100 corrected by 40bit 10MHz counter of 11.797163400 obtained at 1Hz counter of 10. Result: 10.338172850
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-10T04:43:35.338.030.725
   Max 1st sample MIU time among all SL packets: 2021-09-10T04:44:13.208.477.725
   Min 1st sample XO time among all SL packets: 10.338032574
   Max 1st sample XO time among all SL packets: 48.208479513
   TM values from -186 to 182 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-10T04:43:35.338.030.725 (XO: 10.338032574 - 10.338338575, t=37.2 degC)  Rank: 00120
   TM values: Max =   182  Min =  -125  Ave =    -4.19  Std =    24.97
   Min TM value:-125 at an offset of 113.225 usec -> 2021-09-10T04:43:35.338.143.950 (XO: 10.338145800)
   Max TM value:182 at an offset of 107.750 usec -> 2021-09-10T04:43:35.338.138.475 (XO: 10.338140325)
   Max dTM/dt [1/us]:5.18e+02 at an offset of 271.788 usec -> 2021-09-10T04:43:35.338.302.513 (XO: 10.338304363)
   Min dTM/dt [1/us]:-4.80e+02 at an offset of 156.876 usec -> 2021-09-10T04:43:35.338.187.600 (XO: 10.338189450)

 - Block number: 2 starting at 2021-09-10T04:43:52.968.487.725 (XO: 27.968491221 - 27.968797222, t=37.2 degC)  Rank: 00057
   TM values: Max =   147  Min =  -164  Ave =    -4.40  Std =    14.54
   Min TM value:-164 at an offset of 119.550 usec -> 2021-09-10T04:43:52.968.607.275 (XO: 27.968610771)
   Max TM value:147 at an offset of 122.163 usec -> 2021-09-10T04:43:52.968.609.887 (XO: 27.968613384)
   Max dTM/dt [1/us]:1.22e+03 at an offset of 122.025 usec -> 2021-09-10T04:43:52.968.609.750 (XO: 27.968613246)
   Min dTM/dt [1/us]:-1.22e+03 at an offset of 119.350 usec -> 2021-09-10T04:43:52.968.607.075 (XO: 27.968610571)

 - Block number: 3 starting at 2021-09-10T04:44:04.163.470.725 (XO: 39.163472481 - 39.163778482, t=37.2 degC)  Rank: 00108
   TM values: Max =   141  Min =  -171  Ave =    -4.77  Std =    24.64
   Min TM value:-171 at an offset of 102.850 usec -> 2021-09-10T04:44:04.163.573.575 (XO: 39.163575331)
   Max TM value:141 at an offset of 97.738 usec -> 2021-09-10T04:44:04.163.568.462 (XO: 39.163570219)
   Max dTM/dt [1/us]:5.28e+02 at an offset of 228.638 usec -> 2021-09-10T04:44:04.163.699.363 (XO: 39.163701119)
   Min dTM/dt [1/us]:-4.94e+02 at an offset of 135.775 usec -> 2021-09-10T04:44:04.163.606.500 (XO: 39.163608256)

 - Block number: 4 starting at 2021-09-10T04:44:13.208.188.725 (XO: 48.208190512 - 48.208496513, t=37.2 degC)  Rank: 00054
   TM values: Max =   143  Min =  -186  Ave =    -4.56  Std =    13.08
   Min TM value:-186 at an offset of 119.925 usec -> 2021-09-10T04:44:13.208.308.650 (XO: 48.208310437)
   Max TM value:143 at an offset of 119.650 usec -> 2021-09-10T04:44:13.208.308.375 (XO: 48.208310162)
   Max dTM/dt [1/us]:1.32e+03 at an offset of 122.375 usec -> 2021-09-10T04:44:13.208.311.100 (XO: 48.208312887)
   Min dTM/dt [1/us]:-1.54e+03 at an offset of 119.750 usec -> 2021-09-10T04:44:13.208.308.475 (XO: 48.208310262)
