-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan  8 08:52:20 2022 UTC
-- File 20210909_162541_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.095197 to 64.360721
-each separately from 6s to 64s and from 95197us to 693677us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.589518 in 1 packets: 18
      24.008095 in 1 packets: 36
      6.488710 in 1 packets: 54
      18.177756 in 1 packets: 72
MIU packet time from 2021-09-09T16:25:50.095.197.000 to 2021-09-09T16:26:48.360.721.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.589518 in 1 packets: 18
      24.008095 in 1 packets: 36
      6.488710 in 1 packets: 54
      18.177756 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.8398921 to 65.8396977
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999698 in 1 packets: 18
      23.9999195 in 1 packets: 36
      6.9999766 in 1 packets: 54
      17.9999397 in 1 packets: 72
Temperature FPGA: 
      05 -> 48.5 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 36 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, ...
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 53 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 53 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20210909_162541_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.9350903 to 7.9353793
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.5894870  -- OK for SL --
   MIU time from 2021-09-09T16:25:50.095.197.000 to 2021-09-09T16:25:50.095.486.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.589518

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.5248663 to 17.5251553
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.0080150  -- OK for SL --
   MIU time from 2021-09-09T16:25:59.685.004.000 to 2021-09-09T16:25:59.685.293.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.008095

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.5331703 to 41.5334593
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.4886870  -- OK for SL --
   MIU time from 2021-09-09T16:26:23.693.388.000 to 2021-09-09T16:26:23.693.677.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.488710

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 48.0221463 to 48.0224353
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.1776950  -- OK for SL --
   MIU time from 2021-09-09T16:26:30.182.387.000 to 2021-09-09T16:26:30.182.676.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.177756

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 66.2001303 to 66.2004193
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-09T16:26:48.360.432.000 to 2021-09-09T16:26:48.360.721.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 7.935090300 corrected by 40bit 10MHz counter of 7.839892100 obtained at 1Hz counter of 6. Result: 6.095198524
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-09T16:25:50.095.056.725
   Max 1st sample MIU time among all SL packets: 2021-09-09T16:26:48.360.580.725
   Min 1st sample XO time among all SL packets: 6.095058248
   Max 1st sample XO time among all SL packets: 64.360582515
   TM values from -275 to 235 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-09T16:25:50.095.056.725 (XO: 6.095058248 - 6.095364249, t=37.2 degC)  Rank: 00080
   TM values: Max =   235  Min =  -275  Ave =    -4.42  Std =    23.78
   Min TM value:-275 at an offset of 127.875 usec -> 2021-09-09T16:25:50.095.184.600 (XO: 6.095186124)
   Max TM value:235 at an offset of 130.575 usec -> 2021-09-09T16:25:50.095.187.300 (XO: 6.095188824)
   Max dTM/dt [1/us]:2.09e+03 at an offset of 130.450 usec -> 2021-09-09T16:25:50.095.187.175 (XO: 6.095188699)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 127.750 usec -> 2021-09-09T16:25:50.095.184.475 (XO: 6.095185999)

 - Block number: 2 starting at 2021-09-09T16:25:59.684.863.725 (XO: 15.684866385 - 15.685172386, t=37.2 degC)  Rank: 00080
   TM values: Max =   216  Min =  -240  Ave =    -4.60  Std =    22.68
   Min TM value:-240 at an offset of 110.188 usec -> 2021-09-09T16:25:59.684.973.912 (XO: 15.684976573)
   Max TM value:216 at an offset of 125.700 usec -> 2021-09-09T16:25:59.684.989.425 (XO: 15.684992085)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 125.600 usec -> 2021-09-09T16:25:59.684.989.325 (XO: 15.684991985)
   Min dTM/dt [1/us]:-1.97e+03 at an offset of 122.875 usec -> 2021-09-09T16:25:59.684.986.600 (XO: 15.684989260)

 - Block number: 3 starting at 2021-09-09T16:26:23.693.247.725 (XO: 39.693250913 - 39.693556914, t=37.2 degC)  Rank: 00081
   TM values: Max =   226  Min =  -248  Ave =    -4.51  Std =    18.84
   Min TM value:-248 at an offset of 127.900 usec -> 2021-09-09T16:26:23.693.375.625 (XO: 39.693378813)
   Max TM value:226 at an offset of 125.675 usec -> 2021-09-09T16:26:23.693.373.400 (XO: 39.693376588)
   Max dTM/dt [1/us]:2.05e+03 at an offset of 125.538 usec -> 2021-09-09T16:26:23.693.373.262 (XO: 39.693376451)
   Min dTM/dt [1/us]:-2.11e+03 at an offset of 127.700 usec -> 2021-09-09T16:26:23.693.375.425 (XO: 39.693378613)

 - Block number: 4 starting at 2021-09-09T16:26:30.182.246.725 (XO: 46.182248645 - 46.182554646, t=37.2 degC)  Rank: 00078
   TM values: Max =   219  Min =  -275  Ave =    -4.57  Std =    31.96
   Min TM value:-275 at an offset of 115.350 usec -> 2021-09-09T16:26:30.182.362.075 (XO: 46.182363995)
   Max TM value:219 at an offset of 113.075 usec -> 2021-09-09T16:26:30.182.359.800 (XO: 46.182361720)
   Max dTM/dt [1/us]:2.14e+03 at an offset of 117.725 usec -> 2021-09-09T16:26:30.182.364.450 (XO: 46.182366370)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 87.413 usec -> 2021-09-09T16:26:30.182.334.137 (XO: 46.182336057)

 - Block number: 5 starting at 2021-09-09T16:26:48.360.291.725 (XO: 64.360293514 - 64.360599515, t=37.2 degC)  Rank: 00080
   TM values: Max =   226  Min =  -269  Ave =    -4.41  Std =    24.11
   Min TM value:-269 at an offset of 128.413 usec -> 2021-09-09T16:26:48.360.420.137 (XO: 64.360421927)
   Max TM value:226 at an offset of 128.113 usec -> 2021-09-09T16:26:48.360.419.837 (XO: 64.360421627)
   Max dTM/dt [1/us]:2.23e+03 at an offset of 130.763 usec -> 2021-09-09T16:26:48.360.422.487 (XO: 64.360424277)
   Min dTM/dt [1/us]:-2.28e+03 at an offset of 128.200 usec -> 2021-09-09T16:26:48.360.419.925 (XO: 64.360421714)
