-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan  8 06:36:20 2022 UTC
-- File 20210909_144338_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.425082 to 28.196683
-each separately from 4s to 28s and from 196394us to 425371us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.771023 in 1 packets: 18
MIU packet time from 2021-09-09T14:43:45.425.082.000 to 2021-09-09T14:44:09.196.683.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.771023 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.8664490 to 29.8663693
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.9999203 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210909_144338_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.2915318 to 6.2918208
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.7709430  -- OK for SL --
   MIU time from 2021-09-09T14:43:45.425.082.000 to 2021-09-09T14:43:45.425.371.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.771023

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.0627638 to 30.0630528
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-09T14:44:09.196.394.000 to 2021-09-09T14:44:09.196.683.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 6.291531800 corrected by 40bit 10MHz counter of 5.866449000 obtained at 1Hz counter of 4. Result: 4.425084160
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996800
   Min 1st sample MIU time among all SL packets: 2021-09-09T14:43:45.424.941.725
   Max 1st sample MIU time among all SL packets: 2021-09-09T14:44:09.196.542.725
   Min 1st sample XO time among all SL packets: 4.424943885
   Max 1st sample XO time among all SL packets: 28.196543893
   TM values from -264 to 248 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-09T14:43:45.424.941.725 (XO: 4.424943885 - 4.425249886, t=37.2 degC)  Rank: 00080
   TM values: Max =   248  Min =  -264  Ave =    -4.36  Std =    26.66
   Min TM value:-264 at an offset of 200.301 usec -> 2021-09-09T14:43:45.425.142.025 (XO: 4.425144185)
   Max TM value:248 at an offset of 128.900 usec -> 2021-09-09T14:43:45.425.070.625 (XO: 4.425072785)
   Max dTM/dt [1/us]:2.02e+03 at an offset of 128.763 usec -> 2021-09-09T14:43:45.425.070.487 (XO: 4.425072648)
   Min dTM/dt [1/us]:-2.07e+03 at an offset of 200.188 usec -> 2021-09-09T14:43:45.425.141.913 (XO: 4.425144073)

 - Block number: 2 starting at 2021-09-09T14:44:09.196.253.725 (XO: 28.196254892 - 28.196560893, t=37.2 degC)  Rank: 00085
   TM values: Max =   215  Min =  -221  Ave =    -4.40  Std =    20.51
   Min TM value:-221 at an offset of 127.975 usec -> 2021-09-09T14:44:09.196.381.700 (XO: 28.196382868)
   Max TM value:215 at an offset of 130.763 usec -> 2021-09-09T14:44:09.196.384.487 (XO: 28.196385655)
   Max dTM/dt [1/us]:1.91e+03 at an offset of 130.600 usec -> 2021-09-09T14:44:09.196.384.325 (XO: 28.196385493)
   Min dTM/dt [1/us]:-1.65e+03 at an offset of 127.788 usec -> 2021-09-09T14:44:09.196.381.512 (XO: 28.196382680)
