-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan  7 22:14:32 2022 UTC
-- File 20210909_061440_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.139916 to 45.224355
-each separately from 0s to 45s and from 113598us to 252127us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.111633 in 1 packets: 18
      18.861471 in 1 packets: 36
      6.110179 in 1 packets: 54
MIU packet time from 2021-09-09T06:14:43.139.916.000 to 2021-09-09T06:15:28.224.355.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.111633 in 1 packets: 18
      18.861471 in 1 packets: 36
      6.110179 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.9216305 to 46.9214802
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.9999332 in 1 packets: 18
      18.9999365 in 1 packets: 36
      5.9999800 in 1 packets: 54
Temperature FPGA: 
      05 -> 48.5 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210909_061440_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      66 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.0615471 to 2.0618361
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.1115670  -- OK for SL --
   MIU time from 2021-09-09T06:14:43.139.916.000 to 2021-09-09T06:14:43.140.205.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.111633

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      68 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.1734031 to 22.1736921
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.8614070  -- OK for SL --
   MIU time from 2021-09-09T06:15:03.251.838.000 to 2021-09-09T06:15:03.252.127.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.861471

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.0350991 to 41.0353881
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.1101590  -- OK for SL --
   MIU time from 2021-09-09T06:15:22.113.598.000 to 2021-09-09T06:15:22.113.887.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.110179

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 47.1455471 to 47.1458361
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-09T06:15:28.224.066.000 to 2021-09-09T06:15:28.224.355.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.061547100 corrected by 40bit 10MHz counter of 1.921630500 obtained at 1Hz counter of 0. Result: 0.139917076
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-09T06:14:43.139.775.725
   Max 1st sample MIU time among all SL packets: 2021-09-09T06:15:28.224.214.725
   Min 1st sample XO time among all SL packets: 0.139776800
   Max 1st sample XO time among all SL packets: 45.224216365
   TM values from -316 to 276 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-09T06:14:43.139.775.725 (XO: 0.139776800 - 0.140082801, t=37.2 degC)  Rank: 00066
   TM values: Max =   274  Min =  -306  Ave =    -4.46  Std =    34.33
   Min TM value:-306 at an offset of 125.225 usec -> 2021-09-09T06:14:43.139.900.950 (XO: 0.139902026)
   Max TM value:274 at an offset of 127.813 usec -> 2021-09-09T06:14:43.139.903.537 (XO: 0.139904613)
   Max dTM/dt [1/us]:2.52e+03 at an offset of 100.663 usec -> 2021-09-09T06:14:43.139.876.387 (XO: 0.139877463)
   Min dTM/dt [1/us]:-2.45e+03 at an offset of 125.100 usec -> 2021-09-09T06:14:43.139.900.825 (XO: 0.139901901)

 - Block number: 2 starting at 2021-09-09T06:15:03.251.697.725 (XO: 20.251699956 - 20.252005957, t=37.2 degC)  Rank: 00068
   TM values: Max =   276  Min =  -316  Ave =    -4.56  Std =    40.55
   Min TM value:-316 at an offset of 195.663 usec -> 2021-09-09T06:15:03.251.893.388 (XO: 20.251895619)
   Max TM value:276 at an offset of 127.413 usec -> 2021-09-09T06:15:03.251.825.137 (XO: 20.251827369)
   Max dTM/dt [1/us]:2.44e+03 at an offset of 127.288 usec -> 2021-09-09T06:15:03.251.825.012 (XO: 20.251827244)
   Min dTM/dt [1/us]:-2.69e+03 at an offset of 195.538 usec -> 2021-09-09T06:15:03.251.893.263 (XO: 20.251895494)

 - Block number: 3 starting at 2021-09-09T06:15:22.113.457.725 (XO: 39.113458999 - 39.113765000, t=37.2 degC)  Rank: 00078
   TM values: Max =   244  Min =  -285  Ave =    -4.42  Std =    28.11
   Min TM value:-285 at an offset of 230.851 usec -> 2021-09-09T06:15:22.113.688.575 (XO: 39.113689850)
   Max TM value:244 at an offset of 128.463 usec -> 2021-09-09T06:15:22.113.586.187 (XO: 39.113587462)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 128.338 usec -> 2021-09-09T06:15:22.113.586.062 (XO: 39.113587337)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 230.701 usec -> 2021-09-09T06:15:22.113.688.425 (XO: 39.113689700)

 - Block number: 4 starting at 2021-09-09T06:15:28.223.925.725 (XO: 45.223927364 - 45.224233365, t=37.2 degC)  Rank: 00085
   TM values: Max =   232  Min =  -280  Ave =    -4.53  Std =    21.32
   Min TM value:-280 at an offset of 126.925 usec -> 2021-09-09T06:15:28.224.052.650 (XO: 45.224054289)
   Max TM value:232 at an offset of 129.550 usec -> 2021-09-09T06:15:28.224.055.275 (XO: 45.224056914)
   Max dTM/dt [1/us]:1.56e+03 at an offset of 129.400 usec -> 2021-09-09T06:15:28.224.055.125 (XO: 45.224056764)
   Min dTM/dt [1/us]:-2.29e+03 at an offset of 126.800 usec -> 2021-09-09T06:15:28.224.052.525 (XO: 45.224054164)
