-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan  7 13:01:22 2022 UTC
-- File 20210908_153545_1227.dat, 18  packets found.
APID:
      1227 (SL) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.849905 to 10.850194
-each separately from 10s to 10s and from 849905us to 850194us.
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU packet time from 2021-09-08T15:35:58.849.905.000 to 2021-09-08T15:35:58.850.194.000
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.8361671 to 11.8361671
   Differences in neigboring packets: 
      0.0000000 in 17 packets (all packets)
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 18 packets (all packets)
Number of generated alerts: 
      0 in 18 packets (all packets)
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20210908_153545_1227.dat, 1 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.6860721 to 12.6863611
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-08T15:35:58.849.905.000 to 2021-09-08T15:35:58.850.194.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.686072100 corrected by 40bit 10MHz counter of 11.836167100 obtained at 1Hz counter of 10. Result: 10.849907805
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-08T15:35:58.849.764.725
   Max 1st sample MIU time among all SL packets: 2021-09-08T15:35:58.850.053.725
   Min 1st sample XO time among all SL packets: 10.849767529
   Max 1st sample XO time among all SL packets: 10.850056530
   TM values from -251 to 263 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-08T15:35:58.849.764.725 (XO: 10.849767529 - 10.850073530, t=37.2 degC)  Rank: 00079
   TM values: Max =   263  Min =  -251  Ave =    -4.54  Std =    20.44
   Min TM value:-251 at an offset of 128.338 usec -> 2021-09-08T15:35:58.849.893.062 (XO: 10.849895867)
   Max TM value:263 at an offset of 131.050 usec -> 2021-09-08T15:35:58.849.895.775 (XO: 10.849898580)
   Max dTM/dt [1/us]:2.14e+03 at an offset of 126.075 usec -> 2021-09-08T15:35:58.849.890.800 (XO: 10.849893605)
   Min dTM/dt [1/us]:-2.43e+03 at an offset of 128.213 usec -> 2021-09-08T15:35:58.849.892.937 (XO: 10.849895742)
