-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Jan  4 12:15:52 2022 UTC
-- File 20210906_173206_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.639992 to 61.793434
-each separately from 3s to 61s and from 122449us to 793434us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.780603 in 1 packets: 18
      14.290606 in 1 packets: 36
      13.410381 in 1 packets: 54
      12.670407 in 1 packets: 72
MIU packet time from 2021-09-06T17:32:12.639.992.000 to 2021-09-06T17:33:10.793.434.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.780603 in 1 packets: 18
      14.290606 in 1 packets: 36
      13.410381 in 1 packets: 54
      12.670407 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.9228048 to 62.9226118
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.9999401 in 1 packets: 18
      13.9999534 in 2 packets: 36, 54
      11.9999601 in 1 packets: 72
Temperature FPGA: 
      05 -> 48.5 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 36 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 1 packets: 37
      2 in 53 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 1 packets: 37
      2 in 53 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20210906_173206_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.5627976 to 5.5630866
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.7805430  -- OK for SL --
   MIU time from 2021-09-06T17:32:12.639.992.000 to 2021-09-06T17:32:12.640.281.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.780603

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.3436296 to 23.3439186
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.2905590  -- OK for SL --
   MIU time from 2021-09-06T17:32:30.420.884.000 to 2021-09-06T17:32:30.421.173.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.290606

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.6344776 to 37.6347666
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.4103350  -- OK for SL --
   MIU time from 2021-09-06T17:32:44.711.779.000 to 2021-09-06T17:32:44.712.068.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.410381

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 51.0451016 to 51.0453906
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.6703670  -- OK for SL --
   MIU time from 2021-09-06T17:32:58.122.449.000 to 2021-09-06T17:32:58.122.738.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.670407

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 63.7157576 to 63.7160466
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-06T17:33:10.793.145.000 to 2021-09-06T17:33:10.793.434.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.562797600 corrected by 40bit 10MHz counter of 4.922804800 obtained at 1Hz counter of 3. Result: 3.639994912
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-06T17:32:12.639.851.725
   Max 1st sample MIU time among all SL packets: 2021-09-06T17:33:10.793.293.725
   Min 1st sample XO time among all SL packets: 3.639854637
   Max 1st sample XO time among all SL packets: 61.793297143
   TM values from -266 to 254 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-06T17:32:12.639.851.725 (XO: 3.639854637 - 3.640160638, t=37.2 degC)  Rank: 00078
   TM values: Max =   254  Min =  -253  Ave =    -4.55  Std =    26.26
   Min TM value:-253 at an offset of 130.975 usec -> 2021-09-06T17:32:12.639.982.700 (XO: 3.639985612)
   Max TM value:254 at an offset of 128.725 usec -> 2021-09-06T17:32:12.639.980.450 (XO: 3.639983362)
   Max dTM/dt [1/us]:2.36e+03 at an offset of 128.588 usec -> 2021-09-06T17:32:12.639.980.312 (XO: 3.639983224)
   Min dTM/dt [1/us]:-2.35e+03 at an offset of 130.775 usec -> 2021-09-06T17:32:12.639.982.500 (XO: 3.639985412)

 - Block number: 2 starting at 2021-09-06T17:32:30.420.743.725 (XO: 21.420745813 - 21.421051814, t=37.2 degC)  Rank: 00075
   TM values: Max =   248  Min =  -223  Ave =    -4.60  Std =    23.23
   Min TM value:-223 at an offset of 116.200 usec -> 2021-09-06T17:32:30.420.859.925 (XO: 21.420862014)
   Max TM value:248 at an offset of 118.913 usec -> 2021-09-06T17:32:30.420.862.637 (XO: 21.420864726)
   Max dTM/dt [1/us]:1.85e+03 at an offset of 118.763 usec -> 2021-09-06T17:32:30.420.862.487 (XO: 21.420864576)
   Min dTM/dt [1/us]:-2.02e+03 at an offset of 116.075 usec -> 2021-09-06T17:32:30.420.859.800 (XO: 21.420861889)

 - Block number: 3 starting at 2021-09-06T17:32:44.711.638.725 (XO: 35.711641373 - 35.711947374, t=37.2 degC)  Rank: 00082
   TM values: Max =   206  Min =  -266  Ave =    -4.46  Std =    22.95
   Min TM value:-266 at an offset of 113.225 usec -> 2021-09-06T17:32:44.711.751.950 (XO: 35.711754599)
   Max TM value:206 at an offset of 115.850 usec -> 2021-09-06T17:32:44.711.754.575 (XO: 35.711757224)
   Max dTM/dt [1/us]:1.85e+03 at an offset of 115.713 usec -> 2021-09-06T17:32:44.711.754.437 (XO: 35.711757086)
   Min dTM/dt [1/us]:-2.15e+03 at an offset of 113.063 usec -> 2021-09-06T17:32:44.711.751.787 (XO: 35.711754436)

 - Block number: 4 starting at 2021-09-06T17:32:58.122.308.725 (XO: 49.122310029 - 49.122616030, t=37.2 degC)  Rank: 00079
   TM values: Max =   240  Min =  -231  Ave =    -4.48  Std =    19.86
   Min TM value:-231 at an offset of 125.788 usec -> 2021-09-06T17:32:58.122.434.512 (XO: 49.122435817)
   Max TM value:240 at an offset of 128.350 usec -> 2021-09-06T17:32:58.122.437.075 (XO: 49.122438379)
   Max dTM/dt [1/us]:1.98e+03 at an offset of 128.200 usec -> 2021-09-06T17:32:58.122.436.925 (XO: 49.122438229)
   Min dTM/dt [1/us]:-2.06e+03 at an offset of 125.625 usec -> 2021-09-06T17:32:58.122.434.350 (XO: 49.122435654)

 - Block number: 5 starting at 2021-09-06T17:33:10.793.004.725 (XO: 61.793008142 - 61.793314143, t=37.2 degC)  Rank: 00080
   TM values: Max =   208  Min =  -244  Ave =    -4.43  Std =    20.68
   Min TM value:-244 at an offset of 127.688 usec -> 2021-09-06T17:33:10.793.132.412 (XO: 61.793135830)
   Max TM value:208 at an offset of 127.400 usec -> 2021-09-06T17:33:10.793.132.125 (XO: 61.793135542)
   Max dTM/dt [1/us]:2.03e+03 at an offset of 130.075 usec -> 2021-09-06T17:33:10.793.134.800 (XO: 61.793138217)
   Min dTM/dt [1/us]:-2.40e+03 at an offset of 127.513 usec -> 2021-09-06T17:33:10.793.132.237 (XO: 61.793135655)
