-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Jan  4 10:34:58 2022 UTC
-- File 20210906_160227_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.345038 to 23.353037
-each separately from 14s to 23s and from 345038us to 353037us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.007421 in 1 packets: 18
MIU packet time from 2021-09-06T16:02:44.345.038.000 to 2021-09-06T16:02:53.353.037.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.007421 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.8852205 to 24.8851906
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999701 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210906_160227_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      90 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.2302597 to 16.2305487
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.0073910  -- OK for SL --
   MIU time from 2021-09-06T16:02:44.345.038.000 to 2021-09-06T16:02:44.345.327.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.007421

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.2379397 to 25.2382287
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-06T16:02:53.352.748.000 to 2021-09-06T16:02:53.353.037.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 16.230259700 corrected by 40bit 10MHz counter of 15.885220500 obtained at 1Hz counter of 14. Result: 14.345040339
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-06T16:02:44.344.897.725
   Max 1st sample MIU time among all SL packets: 2021-09-06T16:02:53.352.896.725
   Min 1st sample XO time among all SL packets: 14.344900063
   Max 1st sample XO time among all SL packets: 23.352899025
   TM values from -325 to 252 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-06T16:02:44.344.897.725 (XO: 14.344900063 - 14.345206064, t=37.2 degC)  Rank: 00090
   TM values: Max =   252  Min =  -325  Ave =    -4.36  Std =    32.10
   Min TM value:-325 at an offset of 113.463 usec -> 2021-09-06T16:02:44.345.011.187 (XO: 14.345013526)
   Max TM value:252 at an offset of 113.213 usec -> 2021-09-06T16:02:44.345.010.937 (XO: 14.345013276)
   Max dTM/dt [1/us]:2.34e+03 at an offset of 115.888 usec -> 2021-09-06T16:02:44.345.013.612 (XO: 14.345015951)
   Min dTM/dt [1/us]:-3.04e+03 at an offset of 113.350 usec -> 2021-09-06T16:02:44.345.011.075 (XO: 14.345013414)

 - Block number: 2 starting at 2021-09-06T16:02:53.352.607.725 (XO: 23.352610024 - 23.352916025, t=37.2 degC)  Rank: 00084
   TM values: Max =   252  Min =  -250  Ave =    -4.42  Std =    19.91
   Min TM value:-250 at an offset of 113.100 usec -> 2021-09-06T16:02:53.352.720.825 (XO: 23.352723124)
   Max TM value:252 at an offset of 115.813 usec -> 2021-09-06T16:02:53.352.723.537 (XO: 23.352725837)
   Max dTM/dt [1/us]:2.02e+03 at an offset of 110.875 usec -> 2021-09-06T16:02:53.352.718.600 (XO: 23.352720899)
   Min dTM/dt [1/us]:-2.12e+03 at an offset of 112.963 usec -> 2021-09-06T16:02:53.352.720.687 (XO: 23.352722987)
