-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Jan  4 07:32:08 2022 UTC
-- File 20210906_131356_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.629229 to 19.370022
-each separately from 7s to 19s and from 369733us to 629518us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.740215 in 1 packets: 18
MIU packet time from 2021-09-06T13:14:07.629.229.000 to 2021-09-06T13:14:19.370.022.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.740215 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.3250866 to 21.3250466
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999600 in 1 packets: 18
Temperature FPGA: 
      07 -> 45.9 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210906_131356_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.9543165 to 9.9546055
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.7401750  -- OK for SL --
   MIU time from 2021-09-06T13:14:07.629.229.000 to 2021-09-06T13:14:07.629.518.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.740215

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.6947805 to 21.6950695
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-06T13:14:19.369.733.000 to 2021-09-06T13:14:19.370.022.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.954316500 corrected by 40bit 10MHz counter of 9.325086600 obtained at 1Hz counter of 7. Result: 7.629231976
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-06T13:14:07.629.088.725
   Max 1st sample MIU time among all SL packets: 2021-09-06T13:14:19.369.881.725
   Min 1st sample XO time among all SL packets: 7.629091701
   Max 1st sample XO time among all SL packets: 19.369883846
   TM values from -299 to 246 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-06T13:14:07.629.088.725 (XO: 7.629091701 - 7.629397702, t=35.6 degC)  Rank: 00082
   TM values: Max =   246  Min =  -237  Ave =    -4.32  Std =    19.15
   Min TM value:-237 at an offset of 129.100 usec -> 2021-09-06T13:14:07.629.217.825 (XO: 7.629220801)
   Max TM value:246 at an offset of 131.825 usec -> 2021-09-06T13:14:07.629.220.550 (XO: 7.629223526)
   Max dTM/dt [1/us]:2.16e+03 at an offset of 131.688 usec -> 2021-09-06T13:14:07.629.220.412 (XO: 7.629223389)
   Min dTM/dt [1/us]:-2.07e+03 at an offset of 128.938 usec -> 2021-09-06T13:14:07.629.217.662 (XO: 7.629220639)

 - Block number: 2 starting at 2021-09-06T13:14:19.369.592.725 (XO: 19.369594845 - 19.369900846, t=35.6 degC)  Rank: 00082
   TM values: Max =   244  Min =  -299  Ave =    -4.38  Std =    30.31
   Min TM value:-299 at an offset of 172.513 usec -> 2021-09-06T13:14:19.369.765.238 (XO: 19.369767358)
   Max TM value:244 at an offset of 115.350 usec -> 2021-09-06T13:14:19.369.708.075 (XO: 19.369710195)
   Max dTM/dt [1/us]:2.35e+03 at an offset of 118.025 usec -> 2021-09-06T13:14:19.369.710.750 (XO: 19.369712870)
   Min dTM/dt [1/us]:-2.71e+03 at an offset of 115.463 usec -> 2021-09-06T13:14:19.369.708.187 (XO: 19.369710308)
