-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jan  2 14:10:24 2022 UTC
-- File 20210903_153436_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.684372 to 19.237351
-each separately from 4s to 19s and from 237062us to 684661us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.552401 in 1 packets: 18
MIU packet time from 2021-09-03T15:34:43.684.372.000 to 2021-09-03T15:34:58.237.351.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.552401 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.9115967 to 20.9115469
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999502 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210903_153436_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.5959699 to 6.5962589
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.5523510  -- OK for SL --
   MIU time from 2021-09-03T15:34:43.684.372.000 to 2021-09-03T15:34:43.684.661.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.552401

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.1486099 to 21.1488989
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-03T15:34:58.237.062.000 to 2021-09-03T15:34:58.237.351.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 6.595969900 corrected by 40bit 10MHz counter of 5.911596700 obtained at 1Hz counter of 4. Result: 4.684375458
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-03T15:34:43.684.231.725
   Max 1st sample MIU time among all SL packets: 2021-09-03T15:34:58.237.210.725
   Min 1st sample XO time among all SL packets: 4.684235183
   Max 1st sample XO time among all SL packets: 19.237212508
   TM values from -279 to 233 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-03T15:34:43.684.231.725 (XO: 4.684235183 - 4.684541184, t=37.2 degC)  Rank: 00079
   TM values: Max =   230  Min =  -266  Ave =    -4.58  Std =    21.74
   Min TM value:-266 at an offset of 81.013 usec -> 2021-09-03T15:34:43.684.312.737 (XO: 4.684316196)
   Max TM value:230 at an offset of 83.550 usec -> 2021-09-03T15:34:43.684.315.275 (XO: 4.684318733)
   Max dTM/dt [1/us]:2.55e+03 at an offset of 83.400 usec -> 2021-09-03T15:34:43.684.315.125 (XO: 4.684318583)
   Min dTM/dt [1/us]:-2.52e+03 at an offset of 80.863 usec -> 2021-09-03T15:34:43.684.312.587 (XO: 4.684316046)

 - Block number: 2 starting at 2021-09-03T15:34:58.236.921.725 (XO: 19.236923507 - 19.237229508, t=37.2 degC)  Rank: 00088
   TM values: Max =   233  Min =  -279  Ave =    -4.48  Std =    26.54
   Min TM value:-279 at an offset of 116.988 usec -> 2021-09-03T15:34:58.237.038.712 (XO: 19.237040495)
   Max TM value:233 at an offset of 119.625 usec -> 2021-09-03T15:34:58.237.041.350 (XO: 19.237043132)
   Max dTM/dt [1/us]:1.93e+03 at an offset of 114.800 usec -> 2021-09-03T15:34:58.237.036.525 (XO: 19.237038307)
   Min dTM/dt [1/us]:-2.22e+03 at an offset of 116.875 usec -> 2021-09-03T15:34:58.237.038.600 (XO: 19.237040382)
