-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jan  2 07:47:43 2022 UTC
-- File 20210903_102654_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.540577 to 21.761108
-each separately from 11s to 21s and from 540577us to 761108us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.219953 in 1 packets: 18
MIU packet time from 2021-09-03T10:27:08.540.577.000 to 2021-09-03T10:27:18.761.108.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.219953 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.8710087 to 22.8709754
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999667 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210903_102654_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.4115866 to 13.4118756
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.2199190  -- OK for SL --
   MIU time from 2021-09-03T10:27:08.540.577.000 to 2021-09-03T10:27:08.540.866.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.219953

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.6317946 to 23.6320836
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-03T10:27:18.760.819.000 to 2021-09-03T10:27:18.761.108.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 13.411586600 corrected by 40bit 10MHz counter of 12.871008700 obtained at 1Hz counter of 11. Result: 11.540579738
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-09-03T10:27:08.540.436.725
   Max 1st sample MIU time among all SL packets: 2021-09-03T10:27:18.760.967.725
   Min 1st sample XO time among all SL packets: 11.540439462
   Max 1st sample XO time among all SL packets: 21.760970512
   TM values from -272 to 233 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-03T10:27:08.540.436.725 (XO: 11.540439462 - 11.540745464, t=37.2 degC)  Rank: 00081
   TM values: Max =   220  Min =  -272  Ave =    -4.46  Std =    21.08
   Min TM value:-272 at an offset of 119.075 usec -> 2021-09-03T10:27:08.540.555.800 (XO: 11.540558538)
   Max TM value:220 at an offset of 116.763 usec -> 2021-09-03T10:27:08.540.553.487 (XO: 11.540556225)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 116.638 usec -> 2021-09-03T10:27:08.540.553.362 (XO: 11.540556100)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 118.925 usec -> 2021-09-03T10:27:08.540.555.650 (XO: 11.540558388)

 - Block number: 2 starting at 2021-09-03T10:27:18.760.678.725 (XO: 21.760681511 - 21.760987512, t=37.2 degC)  Rank: 00084
   TM values: Max =   233  Min =  -219  Ave =    -4.50  Std =    19.97
   Min TM value:-219 at an offset of 116.163 usec -> 2021-09-03T10:27:18.760.794.887 (XO: 21.760797674)
   Max TM value:233 at an offset of 118.925 usec -> 2021-09-03T10:27:18.760.797.650 (XO: 21.760800437)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 118.813 usec -> 2021-09-03T10:27:18.760.797.537 (XO: 21.760800324)
   Min dTM/dt [1/us]:-1.65e+03 at an offset of 115.988 usec -> 2021-09-03T10:27:18.760.794.712 (XO: 21.760797499)
