-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jan  2 00:58:48 2022 UTC
-- File 20210903_052948_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.212736 to 38.303877
-each separately from 3s to 38s and from 212736us to 393597us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.180283 in 1 packets: 18
      11.909991 in 1 packets: 36
MIU packet time from 2021-09-03T05:29:54.212.736.000 to 2021-09-03T05:30:29.303.877.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.180283 in 1 packets: 18
      11.909991 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.8834259 to 39.8833101
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.9999240 in 1 packets: 18
      11.9999602 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210903_052948_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.0961629 to 5.0964519
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.1802070  -- OK for SL --
   MIU time from 2021-09-03T05:29:54.212.736.000 to 2021-09-03T05:29:54.213.025.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.180283

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.2766589 to 28.2769479
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.9099510  -- OK for SL --
   MIU time from 2021-09-03T05:30:17.393.308.000 to 2021-09-03T05:30:17.393.597.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.909991

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 40.1868989 to 40.1871879
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-03T05:30:29.303.588.000 to 2021-09-03T05:30:29.303.877.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.096162900 corrected by 40bit 10MHz counter of 4.883425900 obtained at 1Hz counter of 3. Result: 3.212737702
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996800
   Min 1st sample MIU time among all SL packets: 2021-09-03T05:29:54.212.595.725
   Max 1st sample MIU time among all SL packets: 2021-09-03T05:30:29.303.736.725
   Min 1st sample XO time among all SL packets: 3.212597427
   Max 1st sample XO time among all SL packets: 38.303738497
   TM values from -249 to 223 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-03T05:29:54.212.595.725 (XO: 3.212597427 - 3.212903428, t=37.2 degC)  Rank: 00084
   TM values: Max =   207  Min =  -249  Ave =    -4.51  Std =    22.50
   Min TM value:-249 at an offset of 130.800 usec -> 2021-09-03T05:29:54.212.726.525 (XO: 3.212728227)
   Max TM value:207 at an offset of 133.350 usec -> 2021-09-03T05:29:54.212.729.075 (XO: 3.212730777)
   Max dTM/dt [1/us]:2.06e+03 at an offset of 133.163 usec -> 2021-09-03T05:29:54.212.728.887 (XO: 3.212730590)
   Min dTM/dt [1/us]:-2.02e+03 at an offset of 130.600 usec -> 2021-09-03T05:29:54.212.726.325 (XO: 3.212728027)

 - Block number: 2 starting at 2021-09-03T05:30:17.393.167.725 (XO: 26.393170022 - 26.393476023, t=37.2 degC)  Rank: 00077
   TM values: Max =   216  Min =  -224  Ave =    -4.48  Std =    18.00
   Min TM value:-224 at an offset of 128.150 usec -> 2021-09-03T05:30:17.393.295.875 (XO: 26.393298173)
   Max TM value:216 at an offset of 130.813 usec -> 2021-09-03T05:30:17.393.298.537 (XO: 26.393300835)
   Max dTM/dt [1/us]:1.89e+03 at an offset of 130.663 usec -> 2021-09-03T05:30:17.393.298.387 (XO: 26.393300685)
   Min dTM/dt [1/us]:-1.72e+03 at an offset of 127.938 usec -> 2021-09-03T05:30:17.393.295.662 (XO: 26.393297960)

 - Block number: 3 starting at 2021-09-03T05:30:29.303.447.725 (XO: 38.303449496 - 38.303755497, t=37.2 degC)  Rank: 00078
   TM values: Max =   223  Min =  -243  Ave =    -4.47  Std =    21.57
   Min TM value:-243 at an offset of 129.975 usec -> 2021-09-03T05:30:29.303.577.700 (XO: 38.303579471)
   Max TM value:223 at an offset of 132.638 usec -> 2021-09-03T05:30:29.303.580.362 (XO: 38.303582134)
   Max dTM/dt [1/us]:1.73e+03 at an offset of 127.738 usec -> 2021-09-03T05:30:29.303.575.462 (XO: 38.303577234)
   Min dTM/dt [1/us]:-1.99e+03 at an offset of 129.813 usec -> 2021-09-03T05:30:29.303.577.537 (XO: 38.303579309)
