-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jan  2 00:54:56 2022 UTC
-- File 20210903_052745_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.509295 to 35.580589
-each separately from 2s to 35s and from 369629us to 580589us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.860045 in 1 packets: 18
      19.210382 in 1 packets: 36
MIU packet time from 2021-09-03T05:27:50.509.295.000 to 2021-09-03T05:28:23.580.589.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.860045 in 1 packets: 18
      19.210382 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.9351688 to 36.9350597
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999538 in 1 packets: 18
      18.9999371 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210903_052745_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.4444646 to 4.4447536
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.8599990  -- OK for SL --
   MIU time from 2021-09-03T05:27:50.509.295.000 to 2021-09-03T05:27:50.509.584.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.860045

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.3047526 to 18.3050416
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.2103190  -- OK for SL --
   MIU time from 2021-09-03T05:28:04.369.629.000 to 2021-09-03T05:28:04.369.918.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.210382

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.5153606 to 37.5156496
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-03T05:28:23.580.300.000 to 2021-09-03T05:28:23.580.589.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.444464600 corrected by 40bit 10MHz counter of 3.935168800 obtained at 1Hz counter of 2. Result: 2.509297481
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-03T05:27:50.509.154.725
   Max 1st sample MIU time among all SL packets: 2021-09-03T05:28:23.580.448.725
   Min 1st sample XO time among all SL packets: 2.509157205
   Max 1st sample XO time among all SL packets: 35.580451540
   TM values from -268 to 233 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-03T05:27:50.509.154.725 (XO: 2.509157205 - 2.509463206, t=37.2 degC)  Rank: 00082
   TM values: Max =   227  Min =  -257  Ave =    -4.59  Std =    21.61
   Min TM value:-257 at an offset of 106.475 usec -> 2021-09-03T05:27:50.509.261.200 (XO: 2.509263681)
   Max TM value:227 at an offset of 121.463 usec -> 2021-09-03T05:27:50.509.276.187 (XO: 2.509278668)
   Max dTM/dt [1/us]:1.74e+03 at an offset of 121.313 usec -> 2021-09-03T05:27:50.509.276.037 (XO: 2.509278518)
   Min dTM/dt [1/us]:-2.07e+03 at an offset of 118.613 usec -> 2021-09-03T05:27:50.509.273.337 (XO: 2.509275818)

 - Block number: 2 starting at 2021-09-03T05:28:04.369.488.725 (XO: 16.369490944 - 16.369796945, t=37.2 degC)  Rank: 00078
   TM values: Max =   214  Min =  -243  Ave =    -4.58  Std =    18.67
   Min TM value:-243 at an offset of 120.075 usec -> 2021-09-03T05:28:04.369.608.800 (XO: 16.369611020)
   Max TM value:214 at an offset of 122.663 usec -> 2021-09-03T05:28:04.369.611.387 (XO: 16.369613607)
   Max dTM/dt [1/us]:1.64e+03 at an offset of 122.488 usec -> 2021-09-03T05:28:04.369.611.212 (XO: 16.369613432)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 119.875 usec -> 2021-09-03T05:28:04.369.608.600 (XO: 16.369610820)

 - Block number: 3 starting at 2021-09-03T05:28:23.580.159.725 (XO: 35.580162540 - 35.580468541, t=37.2 degC)  Rank: 00086
   TM values: Max =   233  Min =  -268  Ave =    -4.45  Std =    22.50
   Min TM value:-268 at an offset of 94.488 usec -> 2021-09-03T05:28:23.580.254.212 (XO: 35.580257027)
   Max TM value:233 at an offset of 92.388 usec -> 2021-09-03T05:28:23.580.252.112 (XO: 35.580254927)
   Max dTM/dt [1/us]:2.48e+03 at an offset of 92.263 usec -> 2021-09-03T05:28:23.580.251.987 (XO: 35.580254802)
   Min dTM/dt [1/us]:-2.41e+03 at an offset of 94.338 usec -> 2021-09-03T05:28:23.580.254.062 (XO: 35.580256877)
