-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Dec 31 07:12:40 2021 UTC
-- File 20210901_123228_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.030546 to 17.611157
-each separately from 2s to 17s and from 30546us to 611157us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.580033 in 1 packets: 18
MIU packet time from 2021-09-01T12:32:33.030.546.000 to 2021-09-01T12:32:48.611.157.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.580033 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.8752362 to 18.8751863
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999501 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210901_123228_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.9057829 to 3.9060719
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.5799830  -- OK for SL --
   MIU time from 2021-09-01T12:32:33.030.546.000 to 2021-09-01T12:32:33.030.835.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.580033

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.4860549 to 19.4863439
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-01T12:32:48.610.868.000 to 2021-09-01T12:32:48.611.157.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 3.905782900 corrected by 40bit 10MHz counter of 3.875236200 obtained at 1Hz counter of 2. Result: 2.030546804
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-01T12:32:33.030.405.725
   Max 1st sample MIU time among all SL packets: 2021-09-01T12:32:48.611.016.725
   Min 1st sample XO time among all SL packets: 2.030406528
   Max 1st sample XO time among all SL packets: 17.611019341
   TM values from -248 to 203 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-01T12:32:33.030.405.725 (XO: 2.030406528 - 2.030712529, t=37.2 degC)  Rank: 00083
   TM values: Max =   203  Min =  -248  Ave =    -4.53  Std =    24.25
   Min TM value:-248 at an offset of 120.825 usec -> 2021-09-01T12:32:33.030.526.550 (XO: 2.030527354)
   Max TM value:203 at an offset of 123.413 usec -> 2021-09-01T12:32:33.030.529.137 (XO: 2.030529941)
   Max dTM/dt [1/us]:1.80e+03 at an offset of 123.225 usec -> 2021-09-01T12:32:33.030.528.950 (XO: 2.030529754)
   Min dTM/dt [1/us]:-1.76e+03 at an offset of 120.613 usec -> 2021-09-01T12:32:33.030.526.337 (XO: 2.030527141)

 - Block number: 2 starting at 2021-09-01T12:32:48.610.727.725 (XO: 17.610730340 - 17.611036341, t=37.2 degC)  Rank: 00081
   TM values: Max =   192  Min =  -215  Ave =    -4.52  Std =    18.43
   Min TM value:-215 at an offset of 109.063 usec -> 2021-09-01T12:32:48.610.836.787 (XO: 17.610839403)
   Max TM value:192 at an offset of 111.763 usec -> 2021-09-01T12:32:48.610.839.487 (XO: 17.610842103)
   Max dTM/dt [1/us]:1.72e+03 at an offset of 111.625 usec -> 2021-09-01T12:32:48.610.839.350 (XO: 17.610841966)
   Min dTM/dt [1/us]:-1.69e+03 at an offset of 108.863 usec -> 2021-09-01T12:32:48.610.836.587 (XO: 17.610839203)
