-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Dec 31 05:49:08 2021 UTC
-- File 20210901_114652_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.844626 to 34.636432
-each separately from 6s to 34s and from 636143us to 844915us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.825991 in 1 packets: 18
      15.964948 in 1 packets: 36
MIU packet time from 2021-09-01T11:47:01.844.626.000 to 2021-09-01T11:47:29.636.432.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.825991 in 1 packets: 18
      15.964948 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.9192556 to 35.9191626
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999601 in 1 packets: 18
      15.9999469 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      C1 -> 37.2 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210901_114652_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.7638824 to 8.7641714
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.8259510  -- OK for SL --
   MIU time from 2021-09-01T11:47:01.844.626.000 to 2021-09-01T11:47:01.844.915.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.825991

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.5901224 to 20.5904114
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.9648950  -- OK for SL --
   MIU time from 2021-09-01T11:47:13.670.906.000 to 2021-09-01T11:47:13.671.195.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.964948

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.5553064 to 36.5555954
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-01T11:47:29.636.143.000 to 2021-09-01T11:47:29.636.432.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.763882400 corrected by 40bit 10MHz counter of 7.919255600 obtained at 1Hz counter of 6. Result: 6.844629587
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-01T11:47:01.844.485.725
   Max 1st sample MIU time among all SL packets: 2021-09-01T11:47:29.636.291.725
   Min 1st sample XO time among all SL packets: 6.844489312
   Max 1st sample XO time among all SL packets: 34.636294625
   TM values from -255 to 218 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-01T11:47:01.844.485.725 (XO: 6.844489312 - 6.844795313, t=35.6 degC)  Rank: 00076
   TM values: Max =   199  Min =  -241  Ave =    -4.51  Std =    19.35
   Min TM value:-241 at an offset of 127.350 usec -> 2021-09-01T11:47:01.844.613.075 (XO: 6.844616662)
   Max TM value:199 at an offset of 127.075 usec -> 2021-09-01T11:47:01.844.612.800 (XO: 6.844616387)
   Max dTM/dt [1/us]:2.20e+03 at an offset of 129.750 usec -> 2021-09-01T11:47:01.844.615.475 (XO: 6.844619062)
   Min dTM/dt [1/us]:-2.29e+03 at an offset of 127.188 usec -> 2021-09-01T11:47:01.844.612.912 (XO: 6.844616500)

 - Block number: 2 starting at 2021-09-01T11:47:13.670.765.725 (XO: 18.670768639 - 18.671074640, t=37.2 degC)  Rank: 00079
   TM values: Max =   196  Min =  -252  Ave =    -4.55  Std =    20.05
   Min TM value:-252 at an offset of 126.338 usec -> 2021-09-01T11:47:13.670.892.062 (XO: 18.670894976)
   Max TM value:196 at an offset of 128.938 usec -> 2021-09-01T11:47:13.670.894.662 (XO: 18.670897576)
   Max dTM/dt [1/us]:1.62e+03 at an offset of 128.763 usec -> 2021-09-01T11:47:13.670.894.487 (XO: 18.670897401)
   Min dTM/dt [1/us]:-1.79e+03 at an offset of 126.200 usec -> 2021-09-01T11:47:13.670.891.925 (XO: 18.670894839)

 - Block number: 3 starting at 2021-09-01T11:47:29.636.002.725 (XO: 34.636005624 - 34.636311625, t=37.2 degC)  Rank: 00081
   TM values: Max =   218  Min =  -255  Ave =    -4.52  Std =    26.51
   Min TM value:-255 at an offset of 197.513 usec -> 2021-09-01T11:47:29.636.200.238 (XO: 34.636203137)
   Max TM value:218 at an offset of 126.488 usec -> 2021-09-01T11:47:29.636.129.212 (XO: 34.636132112)
   Max dTM/dt [1/us]:2.18e+03 at an offset of 129.125 usec -> 2021-09-01T11:47:29.636.131.850 (XO: 34.636134749)
   Min dTM/dt [1/us]:-2.58e+03 at an offset of 126.588 usec -> 2021-09-01T11:47:29.636.129.312 (XO: 34.636132212)
