-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Dec 30 23:51:44 2021 UTC
-- File 20210831_155413_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.223678 to 25.623490
-each separately from 0s to 25s and from 223678us to 623490us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.399234 in 1 packets: 18
MIU packet time from 2021-08-31T15:54:16.223.678.000 to 2021-08-31T15:54:41.623.490.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.399234 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.8414658 to 26.8413827
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.9999169 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 1 packets: 19
      3 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 1 packets: 19
      3 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210831_155413_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.0651449 to 2.0654339
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.3991510  -- OK for SL --
   MIU time from 2021-08-31T15:54:16.223.678.000 to 2021-08-31T15:54:16.223.967.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.399234

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.4645849 to 27.4648739
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-31T15:54:41.623.201.000 to 2021-08-31T15:54:41.623.490.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.065144900 corrected by 40bit 10MHz counter of 1.841465800 obtained at 1Hz counter of 0. Result: 0.223679838
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-31T15:54:16.223.537.725
   Max 1st sample MIU time among all SL packets: 2021-08-31T15:54:41.623.349.725
   Min 1st sample XO time among all SL packets: 0.223539563
   Max 1st sample XO time among all SL packets: 25.623352982
   TM values from -280 to 224 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-31T15:54:16.223.537.725 (XO: 0.223539563 - 0.223845564, t=37.2 degC)  Rank: 00085
   TM values: Max =   224  Min =  -280  Ave =    -4.57  Std =    19.44
   Min TM value:-280 at an offset of 119.113 usec -> 2021-08-31T15:54:16.223.656.837 (XO: 0.223658676)
   Max TM value:224 at an offset of 118.875 usec -> 2021-08-31T15:54:16.223.656.600 (XO: 0.223658438)
   Max dTM/dt [1/us]:2.22e+03 at an offset of 121.550 usec -> 2021-08-31T15:54:16.223.659.275 (XO: 0.223661113)
   Min dTM/dt [1/us]:-2.74e+03 at an offset of 119.000 usec -> 2021-08-31T15:54:16.223.656.725 (XO: 0.223658563)

 - Block number: 2 starting at 2021-08-31T15:54:41.623.060.725 (XO: 25.623063981 - 25.623369982, t=37.2 degC)  Rank: 00083
   TM values: Max =   216  Min =  -267  Ave =    -4.58  Std =    25.57
   Min TM value:-267 at an offset of 127.138 usec -> 2021-08-31T15:54:41.623.187.862 (XO: 25.623191119)
   Max TM value:216 at an offset of 126.850 usec -> 2021-08-31T15:54:41.623.187.575 (XO: 25.623190832)
   Max dTM/dt [1/us]:2.22e+03 at an offset of 129.513 usec -> 2021-08-31T15:54:41.623.190.237 (XO: 25.623193494)
   Min dTM/dt [1/us]:-2.18e+03 at an offset of 126.938 usec -> 2021-08-31T15:54:41.623.187.662 (XO: 25.623190919)
