-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Dec 30 10:40:31 2021 UTC
-- File 20210831_082905_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.262494 to 30.902875
-each separately from 12s to 30s and from 262494us to 902875us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.149840 in 1 packets: 18
      13.489674 in 1 packets: 36
MIU packet time from 2021-08-31T08:29:20.262.494.000 to 2021-08-31T08:29:38.902.875.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.149840 in 1 packets: 18
      13.489674 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.8281278 to 31.8280678
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.9999833 in 1 packets: 18
      12.9999567 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      03 -> 51.1 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      2 in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      2 in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210831_082905_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.0906227 to 14.0909117
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.1498230  -- OK for SL --
   MIU time from 2021-08-31T08:29:20.262.494.000 to 2021-08-31T08:29:20.262.783.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.149840

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.2407347 to 19.2410237
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.4896310  -- OK for SL --
   MIU time from 2021-08-31T08:29:25.412.623.000 to 2021-08-31T08:29:25.412.912.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.489674

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.7306547 to 32.7309437
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-31T08:29:38.902.586.000 to 2021-08-31T08:29:38.902.875.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.090622700 corrected by 40bit 10MHz counter of 13.828127800 obtained at 1Hz counter of 12. Result: 12.262495792
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-08-31T08:29:20.262.353.725
   Max 1st sample MIU time among all SL packets: 2021-08-31T08:29:38.902.734.725
   Min 1st sample XO time among all SL packets: 12.262355517
   Max 1st sample XO time among all SL packets: 30.902738694
   TM values from -267 to 214 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-31T08:29:20.262.353.725 (XO: 12.262355517 - 12.262661518, t=37.2 degC)  Rank: 00079
   TM values: Max =   209  Min =  -260  Ave =    -4.66  Std =    29.04
   Min TM value:-260 at an offset of 183.263 usec -> 2021-08-31T08:29:20.262.536.988 (XO: 12.262538780)
   Max TM value:209 at an offset of 169.351 usec -> 2021-08-31T08:29:20.262.523.075 (XO: 12.262524868)
   Max dTM/dt [1/us]:1.80e+03 at an offset of 127.588 usec -> 2021-08-31T08:29:20.262.481.312 (XO: 12.262483105)
   Min dTM/dt [1/us]:-2.19e+03 at an offset of 169.463 usec -> 2021-08-31T08:29:20.262.523.188 (XO: 12.262524980)

 - Block number: 2 starting at 2021-08-31T08:29:25.412.482.725 (XO: 17.412484727 - 17.412790728, t=37.2 degC)  Rank: 00082
   TM values: Max =   209  Min =  -267  Ave =    -4.64  Std =    29.09
   Min TM value:-267 at an offset of 111.663 usec -> 2021-08-31T08:29:25.412.594.387 (XO: 17.412596390)
   Max TM value:209 at an offset of 114.225 usec -> 2021-08-31T08:29:25.412.596.950 (XO: 17.412598953)
   Max dTM/dt [1/us]:2.00e+03 at an offset of 114.050 usec -> 2021-08-31T08:29:25.412.596.775 (XO: 17.412598778)
   Min dTM/dt [1/us]:-2.06e+03 at an offset of 111.488 usec -> 2021-08-31T08:29:25.412.594.212 (XO: 17.412596215)

 - Block number: 3 starting at 2021-08-31T08:29:38.902.445.725 (XO: 30.902449693 - 30.902755694, t=37.2 degC)  Rank: 00080
   TM values: Max =   214  Min =  -225  Ave =    -4.72  Std =    28.71
   Min TM value:-225 at an offset of 125.413 usec -> 2021-08-31T08:29:38.902.571.137 (XO: 30.902575106)
   Max TM value:214 at an offset of 194.551 usec -> 2021-08-31T08:29:38.902.640.275 (XO: 30.902644244)
   Max dTM/dt [1/us]:1.96e+03 at an offset of 207.663 usec -> 2021-08-31T08:29:38.902.653.388 (XO: 30.902657357)
   Min dTM/dt [1/us]:-1.78e+03 at an offset of 210.051 usec -> 2021-08-31T08:29:38.902.655.775 (XO: 30.902659744)
