-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Dec 30 03:54:54 2021 UTC
-- File 20210830_150906_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.873017 to 27.224009
-each separately from 8s to 27s and from 223720us to 873306us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.350414 in 1 packets: 18
MIU packet time from 2021-08-30T15:09:17.873.017.000 to 2021-08-30T15:09:36.224.009.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.350414 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.8767448 to 28.8766814
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.9999366 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      C0 -> 35.6 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210830_150906_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.7497619 to 10.7500509
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.3503510  -- OK for SL --
   MIU time from 2021-08-30T15:09:17.873.017.000 to 2021-08-30T15:09:17.873.306.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.350414

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.1004019 to 29.1006909
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-30T15:09:36.223.720.000 to 2021-08-30T15:09:36.224.009.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 10.749761900 corrected by 40bit 10MHz counter of 9.876744800 obtained at 1Hz counter of 8. Result: 8.873020068
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-30T15:09:17.872.876.725
   Max 1st sample MIU time among all SL packets: 2021-08-30T15:09:36.223.868.725
   Min 1st sample XO time among all SL packets: 8.872879793
   Max 1st sample XO time among all SL packets: 27.223869964
   TM values from -260 to 228 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-30T15:09:17.872.876.725 (XO: 8.872879793 - 8.873185794, t=37.2 degC)  Rank: 00080
   TM values: Max =   212  Min =  -228  Ave =    -4.50  Std =    18.05
   Min TM value:-228 at an offset of 127.913 usec -> 2021-08-30T15:09:17.873.004.637 (XO: 8.873007706)
   Max TM value:212 at an offset of 130.700 usec -> 2021-08-30T15:09:17.873.007.425 (XO: 8.873010493)
   Max dTM/dt [1/us]:1.76e+03 at an offset of 130.525 usec -> 2021-08-30T15:09:17.873.007.250 (XO: 8.873010318)
   Min dTM/dt [1/us]:-1.68e+03 at an offset of 116.338 usec -> 2021-08-30T15:09:17.872.993.062 (XO: 8.872996131)

 - Block number: 2 starting at 2021-08-30T15:09:36.223.579.725 (XO: 27.223580963 - 27.223886964, t=35.6 degC)  Rank: 00086
   TM values: Max =   228  Min =  -260  Ave =    -4.49  Std =    26.08
   Min TM value:-260 at an offset of 131.088 usec -> 2021-08-30T15:09:36.223.710.812 (XO: 27.223712051)
   Max TM value:228 at an offset of 128.800 usec -> 2021-08-30T15:09:36.223.708.525 (XO: 27.223709763)
   Max dTM/dt [1/us]:1.94e+03 at an offset of 128.688 usec -> 2021-08-30T15:09:36.223.708.412 (XO: 27.223709651)
   Min dTM/dt [1/us]:-2.03e+03 at an offset of 130.938 usec -> 2021-08-30T15:09:36.223.710.662 (XO: 27.223711901)
