-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Dec 30 02:14:07 2021 UTC
-- File 20210830_143254_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.213196 to 17.673272
-each separately from 9s to 17s and from 213196us to 673272us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.459498 in 1 packets: 18
MIU packet time from 2021-08-30T14:33:06.213.196.000 to 2021-08-30T14:33:14.673.272.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.459498 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.6582488 to 18.6582220
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.9999732 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210830_143254_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.8714459 to 10.8717349
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.4594710  -- OK for SL --
   MIU time from 2021-08-30T14:33:06.213.196.000 to 2021-08-30T14:33:06.213.485.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.459498

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.3312059 to 19.3314949
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-30T14:33:14.672.983.000 to 2021-08-30T14:33:14.673.272.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 10.871445900 corrected by 40bit 10MHz counter of 10.658248800 obtained at 1Hz counter of 9. Result: 9.213197825
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-08-30T14:33:06.213.055.725
   Max 1st sample MIU time among all SL packets: 2021-08-30T14:33:14.673.131.725
   Min 1st sample XO time among all SL packets: 9.213057549
   Max 1st sample XO time among all SL packets: 17.673134914
   TM values from -273 to 227 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-30T14:33:06.213.055.725 (XO: 9.213057549 - 9.213363550, t=37.2 degC)  Rank: 00080
   TM values: Max =   202  Min =  -273  Ave =    -4.43  Std =    23.28
   Min TM value:-273 at an offset of 126.563 usec -> 2021-08-30T14:33:06.213.182.287 (XO: 9.213184112)
   Max TM value:202 at an offset of 124.288 usec -> 2021-08-30T14:33:06.213.180.012 (XO: 9.213181837)
   Max dTM/dt [1/us]:2.36e+03 at an offset of 128.950 usec -> 2021-08-30T14:33:06.213.184.675 (XO: 9.213186500)
   Min dTM/dt [1/us]:-2.30e+03 at an offset of 126.375 usec -> 2021-08-30T14:33:06.213.182.100 (XO: 9.213183925)

 - Block number: 2 starting at 2021-08-30T14:33:14.672.842.725 (XO: 17.672845913 - 17.673151914, t=37.2 degC)  Rank: 00086
   TM values: Max =   227  Min =  -267  Ave =    -4.47  Std =    21.01
   Min TM value:-267 at an offset of 126.863 usec -> 2021-08-30T14:33:14.672.969.587 (XO: 17.672972776)
   Max TM value:227 at an offset of 124.275 usec -> 2021-08-30T14:33:14.672.967.000 (XO: 17.672970188)
   Max dTM/dt [1/us]:2.46e+03 at an offset of 126.950 usec -> 2021-08-30T14:33:14.672.969.675 (XO: 17.672972863)
   Min dTM/dt [1/us]:-2.61e+03 at an offset of 124.400 usec -> 2021-08-30T14:33:14.672.967.125 (XO: 17.672970313)
