-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Dec 29 23:57:09 2021 UTC
-- File 20210830_123647_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.582339 to 17.642834
-each separately from 3s to 17s and from 582339us to 642834us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.059917 in 1 packets: 18
MIU packet time from 2021-08-30T12:36:53.582.339.000 to 2021-08-30T12:37:07.642.834.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.059917 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.8891499 to 18.8891034
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999535 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210830_123647_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.4714892 to 5.4717782
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.0598710  -- OK for SL --
   MIU time from 2021-08-30T12:36:53.582.339.000 to 2021-08-30T12:36:53.582.628.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.059917

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.5316492 to 19.5319382
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-30T12:37:07.642.545.000 to 2021-08-30T12:37:07.642.834.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.471489200 corrected by 40bit 10MHz counter of 4.889149900 obtained at 1Hz counter of 3. Result: 3.582341222
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-30T12:36:53.582.198.725
   Max 1st sample MIU time among all SL packets: 2021-08-30T12:37:07.642.693.725
   Min 1st sample XO time among all SL packets: 3.582200946
   Max 1st sample XO time among all SL packets: 17.642696646
   TM values from -297 to 242 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-30T12:36:53.582.198.725 (XO: 3.582200946 - 3.582506947, t=37.2 degC)  Rank: 00080
   TM values: Max =   229  Min =  -243  Ave =    -4.64  Std =    20.70
   Min TM value:-243 at an offset of 101.638 usec -> 2021-08-30T12:36:53.582.300.362 (XO: 3.582302584)
   Max TM value:229 at an offset of 104.250 usec -> 2021-08-30T12:36:53.582.302.975 (XO: 3.582305197)
   Max dTM/dt [1/us]:1.94e+03 at an offset of 104.050 usec -> 2021-08-30T12:36:53.582.302.775 (XO: 3.582304997)
   Min dTM/dt [1/us]:-2.35e+03 at an offset of 101.463 usec -> 2021-08-30T12:36:53.582.300.187 (XO: 3.582302409)

 - Block number: 2 starting at 2021-08-30T12:37:07.642.404.725 (XO: 17.642407645 - 17.642713646, t=37.2 degC)  Rank: 00084
   TM values: Max =   242  Min =  -297  Ave =    -4.65  Std =    32.07
   Min TM value:-297 at an offset of 105.375 usec -> 2021-08-30T12:37:07.642.510.100 (XO: 17.642513020)
   Max TM value:242 at an offset of 105.138 usec -> 2021-08-30T12:37:07.642.509.862 (XO: 17.642512783)
   Max dTM/dt [1/us]:2.45e+03 at an offset of 107.775 usec -> 2021-08-30T12:37:07.642.512.500 (XO: 17.642515420)
   Min dTM/dt [1/us]:-2.93e+03 at an offset of 105.250 usec -> 2021-08-30T12:37:07.642.509.975 (XO: 17.642512895)
