-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Dec 29 11:26:45 2021 UTC
-- File 20210830_060302_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.814930 to 74.133227
-each separately from 9s to 74s and from 132938us to 815219us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.344679 in 1 packets: 18
      12.649479 in 1 packets: 36
      20.555013 in 1 packets: 54
      14.767681 in 1 packets: 72
MIU packet time from 2021-08-30T06:03:14.814.930.000 to 2021-08-30T06:04:19.133.227.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.344679 in 1 packets: 18
      12.649479 in 1 packets: 36
      20.555013 in 1 packets: 54
      14.767681 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.5688424 to 74.5686254
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999433 in 1 packets: 18
      11.9999599 in 1 packets: 36
      20.9999299 in 1 packets: 54
      14.9999499 in 1 packets: 72
Temperature FPGA: 
      03 -> 51.1 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 36 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, ...
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      1 in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      2 in 54 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 5 packets: 2, 3, 4, 5, 6
      3 in 30 packets: 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, 20, 21, 22, 23, 24, ...
      4 in 1 packets: 37
      5 in 53 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20210830_060302_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      141 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.3837727 to 10.3840617
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.3446230  -- OK for SL --
   MIU time from 2021-08-30T06:03:14.814.930.000 to 2021-08-30T06:03:14.815.219.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.344679

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      62 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.7286847 to 26.7289737
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.6494390  -- OK for SL --
   MIU time from 2021-08-30T06:03:31.159.898.000 to 2021-08-30T06:03:31.160.187.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.649479

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      180 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.3784127 to 39.3787017
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.5549430  -- OK for SL --
   MIU time from 2021-08-30T06:03:43.809.666.000 to 2021-08-30T06:03:43.809.955.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.555013

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      71 in 18 packets (all packets)  -- OK --
   10 MHz counter from 59.9336447 to 59.9339337
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.7676310  -- OK for SL --
   MIU time from 2021-08-30T06:04:04.364.968.000 to 2021-08-30T06:04:04.365.257.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.767681

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      71 in 18 packets (all packets)  -- OK --
   10 MHz counter from 74.7015647 to 74.7018537
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-30T06:04:19.132.938.000 to 2021-08-30T06:04:19.133.227.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 75 XO calibration points at 1Hz counter from 0 to 74
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 10.383772700 corrected by 40bit 10MHz counter of 9.568842400 obtained at 1Hz counter of 9. Result: 9.814932989
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-30T06:03:14.814.789.725
   Max 1st sample MIU time among all SL packets: 2021-08-30T06:04:19.133.086.725
   Min 1st sample XO time among all SL packets: 9.814792714
   Max 1st sample XO time among all SL packets: 74.133088478
   TM values from -1492 to 760 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-30T06:03:14.814.789.725 (XO: 9.814792714 - 9.815098715, t=37.2 degC)  Rank: 00141
   TM values: Max =   447  Min =  -694  Ave =    -4.43  Std =    72.10
   Min TM value:-694 at an offset of 117.288 usec -> 2021-08-30T06:03:14.814.907.012 (XO: 9.814910002)
   Max TM value:447 at an offset of 108.038 usec -> 2021-08-30T06:03:14.814.897.762 (XO: 9.814900752)
   Max dTM/dt [1/us]:6.27e+03 at an offset of 115.188 usec -> 2021-08-30T06:03:14.814.904.912 (XO: 9.814907902)
   Min dTM/dt [1/us]:-6.08e+03 at an offset of 115.100 usec -> 2021-08-30T06:03:14.814.904.825 (XO: 9.814907814)

 - Block number: 2 starting at 2021-08-30T06:03:31.159.757.725 (XO: 26.159759268 - 26.160065269, t=37.2 degC)  Rank: 00062
   TM values: Max =   171  Min =  -211  Ave =    -4.51  Std =    19.12
   Min TM value:-211 at an offset of 89.738 usec -> 2021-08-30T06:03:31.159.847.462 (XO: 26.159849006)
   Max TM value:171 at an offset of 92.275 usec -> 2021-08-30T06:03:31.159.850.000 (XO: 26.159851543)
   Max dTM/dt [1/us]:1.66e+03 at an offset of 92.138 usec -> 2021-08-30T06:03:31.159.849.862 (XO: 26.159851406)
   Min dTM/dt [1/us]:-1.83e+03 at an offset of 89.538 usec -> 2021-08-30T06:03:31.159.847.262 (XO: 26.159848806)

 - Block number: 3 starting at 2021-08-30T06:03:43.809.525.725 (XO: 38.809529577 - 38.809835578, t=37.2 degC)  Rank: 00180
   TM values: Max =   760  Min = -1492  Ave =    -4.47  Std =    82.98
   Min TM value:-1492 at an offset of 106.138 usec -> 2021-08-30T06:03:43.809.631.862 (XO: 38.809635715)
   Max TM value:760 at an offset of 106.338 usec -> 2021-08-30T06:03:43.809.632.062 (XO: 38.809635915)
   Max dTM/dt [1/us]:1.24e+04 at an offset of 106.175 usec -> 2021-08-30T06:03:43.809.631.900 (XO: 38.809635753)
   Min dTM/dt [1/us]:-1.24e+04 at an offset of 106.088 usec -> 2021-08-30T06:03:43.809.631.812 (XO: 38.809635665)

 - Block number: 4 starting at 2021-08-30T06:04:04.364.827.725 (XO: 59.364830129 - 59.365136130, t=37.2 degC)  Rank: 00071
   TM values: Max =   212  Min =  -230  Ave =    -4.42  Std =    19.02
   Min TM value:-230 at an offset of 120.075 usec -> 2021-08-30T06:04:04.364.947.800 (XO: 59.364950204)
   Max TM value:212 at an offset of 122.550 usec -> 2021-08-30T06:04:04.364.950.275 (XO: 59.364952679)
   Max dTM/dt [1/us]:1.72e+03 at an offset of 117.838 usec -> 2021-08-30T06:04:04.364.945.562 (XO: 59.364947967)
   Min dTM/dt [1/us]:-2.09e+03 at an offset of 119.938 usec -> 2021-08-30T06:04:04.364.947.662 (XO: 59.364950067)

 - Block number: 5 starting at 2021-08-30T06:04:19.132.797.725 (XO: 74.132799477 - 74.133105478, t=37.2 degC)  Rank: 00071
   TM values: Max =   211  Min =  -238  Ave =    -4.51  Std =    21.51
   Min TM value:-238 at an offset of 126.500 usec -> 2021-08-30T06:04:19.132.924.225 (XO: 74.132925977)
   Max TM value:211 at an offset of 128.988 usec -> 2021-08-30T06:04:19.132.926.712 (XO: 74.132928464)
   Max dTM/dt [1/us]:1.83e+03 at an offset of 128.850 usec -> 2021-08-30T06:04:19.132.926.575 (XO: 74.132928327)
   Min dTM/dt [1/us]:-2.00e+03 at an offset of 126.275 usec -> 2021-08-30T06:04:19.132.924.000 (XO: 74.132925752)
