-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Dec 27 23:59:09 2021 UTC
-- File 20210828_170435_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.928243 to 28.450842
-each separately from 2s to 28s and from 450553us to 928532us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.522021 in 1 packets: 18
MIU packet time from 2021-08-28T17:04:40.928.243.000 to 2021-08-28T17:05:06.450.842.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.522021 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.7234510 to 29.7233644
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.9999134 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210828_170435_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.6516944 to 4.6519834
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.5219350  -- OK for SL --
   MIU time from 2021-08-28T17:04:40.928.243.000 to 2021-08-28T17:04:40.928.532.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.522021

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.1739184 to 30.1742074
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-28T17:05:06.450.553.000 to 2021-08-28T17:05:06.450.842.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.651694400 corrected by 40bit 10MHz counter of 3.723451000 obtained at 1Hz counter of 2. Result: 2.928246463
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-28T17:04:40.928.102.725
   Max 1st sample MIU time among all SL packets: 2021-08-28T17:05:06.450.701.725
   Min 1st sample XO time among all SL packets: 2.928106188
   Max 1st sample XO time among all SL packets: 28.450704212
   TM values from -280 to 267 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-28T17:04:40.928.102.725 (XO: 2.928106188 - 2.928412189, t=37.2 degC)  Rank: 00081
   TM values: Max =   222  Min =  -259  Ave =    -4.53  Std =    20.89
   Min TM value:-259 at an offset of 115.275 usec -> 2021-08-28T17:04:40.928.218.000 (XO: 2.928221463)
   Max TM value:222 at an offset of 117.875 usec -> 2021-08-28T17:04:40.928.220.600 (XO: 2.928224063)
   Max dTM/dt [1/us]:1.76e+03 at an offset of 117.688 usec -> 2021-08-28T17:04:40.928.220.412 (XO: 2.928223876)
   Min dTM/dt [1/us]:-2.29e+03 at an offset of 115.100 usec -> 2021-08-28T17:04:40.928.217.825 (XO: 2.928221288)

 - Block number: 2 starting at 2021-08-28T17:05:06.450.412.725 (XO: 28.450415211 - 28.450721212, t=37.2 degC)  Rank: 00083
   TM values: Max =   267  Min =  -280  Ave =    -4.48  Std =    31.97
   Min TM value:-280 at an offset of 147.488 usec -> 2021-08-28T17:05:06.450.560.212 (XO: 28.450562699)
   Max TM value:267 at an offset of 122.038 usec -> 2021-08-28T17:05:06.450.534.762 (XO: 28.450537249)
   Max dTM/dt [1/us]:2.76e+03 at an offset of 121.913 usec -> 2021-08-28T17:05:06.450.534.637 (XO: 28.450537124)
   Min dTM/dt [1/us]:-2.72e+03 at an offset of 119.413 usec -> 2021-08-28T17:05:06.450.532.137 (XO: 28.450534624)
