-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Dec 27 15:13:00 2021 UTC
-- File 20210828_132326_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.532206 to 20.431982
-each separately from 1s to 20s and from 431693us to 532495us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.899198 in 1 packets: 18
MIU packet time from 2021-08-28T13:23:30.532.206.000 to 2021-08-28T13:23:49.431.982.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.899198 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.9413266 to 21.9412636
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.9999370 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      3 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      3 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210828_132326_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.4735334 to 3.4738224
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.8991350  -- OK for SL --
   MIU time from 2021-08-28T13:23:30.532.206.000 to 2021-08-28T13:23:30.532.495.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.899198

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.3729574 to 22.3732464
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-28T13:23:49.431.693.000 to 2021-08-28T13:23:49.431.982.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.473533400 corrected by 40bit 10MHz counter of 2.941326600 obtained at 1Hz counter of 1. Result: 1.532208610
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-08-28T13:23:30.532.065.725
   Max 1st sample MIU time among all SL packets: 2021-08-28T13:23:49.431.841.725
   Min 1st sample XO time among all SL packets: 1.532068334
   Max 1st sample XO time among all SL packets: 20.431843993
   TM values from -274 to 225 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-28T13:23:30.532.065.725 (XO: 1.532068334 - 1.532374335, t=37.2 degC)  Rank: 00087
   TM values: Max =   220  Min =  -274  Ave =    -4.52  Std =    20.66
   Min TM value:-274 at an offset of 107.750 usec -> 2021-08-28T13:23:30.532.173.475 (XO: 1.532176084)
   Max TM value:220 at an offset of 107.488 usec -> 2021-08-28T13:23:30.532.173.212 (XO: 1.532175822)
   Max dTM/dt [1/us]:2.18e+03 at an offset of 110.175 usec -> 2021-08-28T13:23:30.532.175.900 (XO: 1.532178509)
   Min dTM/dt [1/us]:-2.54e+03 at an offset of 107.613 usec -> 2021-08-28T13:23:30.532.173.337 (XO: 1.532175947)

 - Block number: 2 starting at 2021-08-28T13:23:49.431.552.725 (XO: 20.431554992 - 20.431860993, t=37.2 degC)  Rank: 00080
   TM values: Max =   225  Min =  -239  Ave =    -4.46  Std =    19.38
   Min TM value:-239 at an offset of 105.138 usec -> 2021-08-28T13:23:49.431.657.862 (XO: 20.431660130)
   Max TM value:225 at an offset of 107.775 usec -> 2021-08-28T13:23:49.431.660.500 (XO: 20.431662768)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 107.638 usec -> 2021-08-28T13:23:49.431.660.362 (XO: 20.431662630)
   Min dTM/dt [1/us]:-1.96e+03 at an offset of 104.938 usec -> 2021-08-28T13:23:49.431.657.662 (XO: 20.431659930)
