-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Dec 27 13:43:52 2021 UTC
-- File 20210828_124407_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.912838 to 23.290533
-each separately from 9s to 23s and from 290244us to 913127us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.377117 in 1 packets: 18
MIU packet time from 2021-08-28T12:44:19.912.838.000 to 2021-08-28T12:44:33.290.533.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.377117 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.9477864 to 24.9477401
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999537 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210828_124407_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.8606251 to 11.8609141
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.3770710  -- OK for SL --
   MIU time from 2021-08-28T12:44:19.912.838.000 to 2021-08-28T12:44:19.913.127.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.377117

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.2379851 to 25.2382741
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-28T12:44:33.290.244.000 to 2021-08-28T12:44:33.290.533.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.860625100 corrected by 40bit 10MHz counter of 10.947786400 obtained at 1Hz counter of 9. Result: 9.912841712
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-28T12:44:19.912.697.725
   Max 1st sample MIU time among all SL packets: 2021-08-28T12:44:33.290.392.725
   Min 1st sample XO time among all SL packets: 9.912701437
   Max 1st sample XO time among all SL packets: 23.290394683
   TM values from -242 to 230 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-28T12:44:19.912.697.725 (XO: 9.912701437 - 9.913007438, t=37.2 degC)  Rank: 00087
   TM values: Max =   230  Min =  -242  Ave =    -4.50  Std =    21.71
   Min TM value:-242 at an offset of 130.225 usec -> 2021-08-28T12:44:19.912.827.950 (XO: 9.912831662)
   Max TM value:230 at an offset of 132.750 usec -> 2021-08-28T12:44:19.912.830.475 (XO: 9.912834187)
   Max dTM/dt [1/us]:2.06e+03 at an offset of 127.913 usec -> 2021-08-28T12:44:19.912.825.637 (XO: 9.912829350)
   Min dTM/dt [1/us]:-2.28e+03 at an offset of 130.063 usec -> 2021-08-28T12:44:19.912.827.787 (XO: 9.912831500)

 - Block number: 2 starting at 2021-08-28T12:44:33.290.103.725 (XO: 23.290105682 - 23.290411683, t=37.2 degC)  Rank: 00085
   TM values: Max =   225  Min =  -242  Ave =    -4.46  Std =    16.79
   Min TM value:-242 at an offset of 44.150 usec -> 2021-08-28T12:44:33.290.147.875 (XO: 23.290149832)
   Max TM value:225 at an offset of 41.950 usec -> 2021-08-28T12:44:33.290.145.675 (XO: 23.290147632)
   Max dTM/dt [1/us]:2.14e+03 at an offset of 41.825 usec -> 2021-08-28T12:44:33.290.145.550 (XO: 23.290147507)
   Min dTM/dt [1/us]:-2.28e+03 at an offset of 43.988 usec -> 2021-08-28T12:44:33.290.147.712 (XO: 23.290149670)
