-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Dec 27 12:51:42 2021 UTC
-- File 20210828_121908_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.575629 to 88.176811
-each separately from 2s to 88s and from 80907us to 575918us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.784558 in 1 packets: 18
      8.877581 in 1 packets: 36
      16.266244 in 1 packets: 54
      17.575739 in 1 packets: 72
      19.095326 in 1 packets: 90
MIU packet time from 2021-08-28T12:19:13.575.629.000 to 2021-08-28T12:20:39.176.811.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.784558 in 1 packets: 18
      8.877581 in 1 packets: 36
      16.266244 in 1 packets: 54
      17.575739 in 1 packets: 72
      19.095326 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.9379258 to 89.9376409
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.9999205 in 1 packets: 18
      8.9999702 in 1 packets: 36
      15.9999469 in 1 packets: 54
      17.9999404 in 1 packets: 72
      18.9999371 in 1 packets: 90
Temperature FPGA: 
      04 -> 49.8 degC in 90 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 73, 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Temperature XO: 
      C1 -> 37.2 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 73 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90, 91
      2 in 17 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108
Number of generated alerts: 
      0 in 73 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90, 91
      2 in 17 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20210828_121908_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.5135555 to 4.5138445
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.7844790  -- OK for SL --
   MIU time from 2021-08-28T12:19:13.575.629.000 to 2021-08-28T12:19:13.575.918.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.784558

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.2983235 to 28.2986125
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.8775510  -- OK for SL --
   MIU time from 2021-08-28T12:19:37.360.476.000 to 2021-08-28T12:19:37.360.765.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.877581

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.1761635 to 37.1764525
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.2661910  -- OK for SL --
   MIU time from 2021-08-28T12:19:46.238.346.000 to 2021-08-28T12:19:46.238.635.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.266244

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 53.4426435 to 53.4429325
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.5756790  -- OK for SL --
   MIU time from 2021-08-28T12:20:02.504.879.000 to 2021-08-28T12:20:02.505.168.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.575739

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 71.0186115 to 71.0189005
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.0952630  -- OK for SL --
   MIU time from 2021-08-28T12:20:20.080.907.000 to 2021-08-28T12:20:20.081.196.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.095326

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 90.1141635 to 90.1144525
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-28T12:20:39.176.522.000 to 2021-08-28T12:20:39.176.811.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.513555500 corrected by 40bit 10MHz counter of 3.937925800 obtained at 1Hz counter of 2. Result: 2.575631600
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-28T12:19:13.575.488.725
   Max 1st sample MIU time among all SL packets: 2021-08-28T12:20:39.176.670.725
   Min 1st sample XO time among all SL packets: 2.575491324
   Max 1st sample XO time among all SL packets: 88.176671908
   TM values from -237 to 214 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-28T12:19:13.575.488.725 (XO: 2.575491324 - 2.575797325, t=37.2 degC)  Rank: 00075
   TM values: Max =   182  Min =  -235  Ave =    -4.48  Std =    25.89
   Min TM value:-235 at an offset of 114.750 usec -> 2021-08-28T12:19:13.575.603.475 (XO: 2.575606075)
   Max TM value:182 at an offset of 112.463 usec -> 2021-08-28T12:19:13.575.601.187 (XO: 2.575603787)
   Max dTM/dt [1/us]:1.48e+03 at an offset of 112.300 usec -> 2021-08-28T12:19:13.575.601.025 (XO: 2.575603624)
   Min dTM/dt [1/us]:-1.76e+03 at an offset of 187.376 usec -> 2021-08-28T12:19:13.575.676.100 (XO: 2.575678700)

 - Block number: 2 starting at 2021-08-28T12:19:37.360.335.725 (XO: 26.360338114 - 26.360644115, t=37.2 degC)  Rank: 00075
   TM values: Max =   180  Min =  -227  Ave =    -4.40  Std =    25.45
   Min TM value:-227 at an offset of 124.863 usec -> 2021-08-28T12:19:37.360.460.587 (XO: 26.360462977)
   Max TM value:180 at an offset of 127.450 usec -> 2021-08-28T12:19:37.360.463.175 (XO: 26.360465565)
   Max dTM/dt [1/us]:1.50e+03 at an offset of 157.713 usec -> 2021-08-28T12:19:37.360.493.438 (XO: 26.360495827)
   Min dTM/dt [1/us]:-1.60e+03 at an offset of 155.076 usec -> 2021-08-28T12:19:37.360.490.800 (XO: 26.360493190)

 - Block number: 3 starting at 2021-08-28T12:19:46.238.205.725 (XO: 35.238207535 - 35.238513536, t=37.2 degC)  Rank: 00077
   TM values: Max =   214  Min =  -235  Ave =    -4.46  Std =    22.11
   Min TM value:-235 at an offset of 112.663 usec -> 2021-08-28T12:19:46.238.318.387 (XO: 35.238320198)
   Max TM value:214 at an offset of 115.175 usec -> 2021-08-28T12:19:46.238.320.900 (XO: 35.238322710)
   Max dTM/dt [1/us]:1.95e+03 at an offset of 110.350 usec -> 2021-08-28T12:19:46.238.316.075 (XO: 35.238317885)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 112.500 usec -> 2021-08-28T12:19:46.238.318.225 (XO: 35.238320035)

 - Block number: 4 starting at 2021-08-28T12:20:02.504.738.725 (XO: 51.504741491 - 51.505047492, t=37.2 degC)  Rank: 00077
   TM values: Max =   208  Min =  -237  Ave =    -4.60  Std =    23.05
   Min TM value:-237 at an offset of 114.450 usec -> 2021-08-28T12:20:02.504.853.175 (XO: 51.504855941)
   Max TM value:208 at an offset of 116.963 usec -> 2021-08-28T12:20:02.504.855.687 (XO: 51.504858454)
   Max dTM/dt [1/us]:2.04e+03 at an offset of 116.813 usec -> 2021-08-28T12:20:02.504.855.537 (XO: 51.504858304)
   Min dTM/dt [1/us]:-1.99e+03 at an offset of 114.250 usec -> 2021-08-28T12:20:02.504.852.975 (XO: 51.504855741)

 - Block number: 5 starting at 2021-08-28T12:20:20.080.766.725 (XO: 69.080767692 - 69.081073693, t=37.2 degC)  Rank: 00083
   TM values: Max =   174  Min =  -232  Ave =    -4.56  Std =    19.13
   Min TM value:-232 at an offset of 112.175 usec -> 2021-08-28T12:20:20.080.878.900 (XO: 69.080879867)
   Max TM value:174 at an offset of 114.750 usec -> 2021-08-28T12:20:20.080.881.475 (XO: 69.080882442)
   Max dTM/dt [1/us]:1.57e+03 at an offset of 114.600 usec -> 2021-08-28T12:20:20.080.881.325 (XO: 69.080882292)
   Min dTM/dt [1/us]:-1.50e+03 at an offset of 111.975 usec -> 2021-08-28T12:20:20.080.878.700 (XO: 69.080879667)

 - Block number: 6 starting at 2021-08-28T12:20:39.176.381.725 (XO: 88.176382907 - 88.176688908, t=37.2 degC)  Rank: 00082
   TM values: Max =   177  Min =  -227  Ave =    -4.45  Std =    18.67
   Min TM value:-227 at an offset of 126.525 usec -> 2021-08-28T12:20:39.176.508.250 (XO: 88.176509432)
   Max TM value:177 at an offset of 126.225 usec -> 2021-08-28T12:20:39.176.507.950 (XO: 88.176509132)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 128.913 usec -> 2021-08-28T12:20:39.176.510.637 (XO: 88.176511820)
   Min dTM/dt [1/us]:-1.93e+03 at an offset of 126.338 usec -> 2021-08-28T12:20:39.176.508.062 (XO: 88.176509245)
