-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Dec 27 09:14:35 2021 UTC
-- File 20210828_071850_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.103738 to 63.868090
-each separately from 10s to 63s and from 103738us to 868090us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.457686 in 1 packets: 18
      25.892197 in 1 packets: 36
      11.286323 in 1 packets: 54
      9.126701 in 1 packets: 72
MIU packet time from 2021-08-28T07:19:03.103.738.000 to 2021-08-28T07:19:56.868.090.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.457686 in 1 packets: 18
      25.892197 in 1 packets: 36
      11.286323 in 1 packets: 54
      9.126701 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.9447354 to 64.9445604
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.9999769 in 1 packets: 18
      25.9999141 in 1 packets: 36
      10.9999637 in 1 packets: 54
      8.9999703 in 1 packets: 72
Temperature FPGA: 
      05 -> 48.5 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      03 -> 51.1 degC in 18 packets: 73, 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 54 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 54 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20210828_071850_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.0484743 to 12.0487633
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.4576630  -- OK for SL --
   MIU time from 2021-08-28T07:19:03.103.738.000 to 2021-08-28T07:19:03.104.027.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.457686

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.5064263 to 19.5067153
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.8921110  -- OK for SL --
   MIU time from 2021-08-28T07:19:10.561.713.000 to 2021-08-28T07:19:10.562.002.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.892197

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 45.3988263 to 45.3991153
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.2862870  -- OK for SL --
   MIU time from 2021-08-28T07:19:36.454.199.000 to 2021-08-28T07:19:36.454.488.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.286323

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 56.6854023 to 56.6856913
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.1266710  -- OK for SL --
   MIU time from 2021-08-28T07:19:47.740.811.000 to 2021-08-28T07:19:47.741.100.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.126701

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 65.8123623 to 65.8126513
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-28T07:19:56.867.801.000 to 2021-08-28T07:19:56.868.090.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.048474300 corrected by 40bit 10MHz counter of 11.944735400 obtained at 1Hz counter of 10. Result: 10.103739242
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-28T07:19:03.103.597.725
   Max 1st sample MIU time among all SL packets: 2021-08-28T07:19:56.867.949.725
   Min 1st sample XO time among all SL packets: 10.103598967
   Max 1st sample XO time among all SL packets: 63.867953489
   TM values from -248 to 201 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-28T07:19:03.103.597.725 (XO: 10.103598967 - 10.103904968, t=37.2 degC)  Rank: 00081
   TM values: Max =   194  Min =  -218  Ave =    -4.47  Std =    17.02
   Min TM value:-218 at an offset of 124.775 usec -> 2021-08-28T07:19:03.103.722.500 (XO: 10.103723742)
   Max TM value:194 at an offset of 127.388 usec -> 2021-08-28T07:19:03.103.725.112 (XO: 10.103726355)
   Max dTM/dt [1/us]:1.54e+03 at an offset of 127.225 usec -> 2021-08-28T07:19:03.103.724.950 (XO: 10.103726192)
   Min dTM/dt [1/us]:-1.80e+03 at an offset of 124.575 usec -> 2021-08-28T07:19:03.103.722.300 (XO: 10.103723542)

 - Block number: 2 starting at 2021-08-28T07:19:10.561.572.725 (XO: 17.561575578 - 17.561881579, t=37.2 degC)  Rank: 00080
   TM values: Max =   195  Min =  -240  Ave =    -4.52  Std =    22.68
   Min TM value:-240 at an offset of 89.238 usec -> 2021-08-28T07:19:10.561.661.962 (XO: 17.561664816)
   Max TM value:195 at an offset of 91.800 usec -> 2021-08-28T07:19:10.561.664.525 (XO: 17.561667379)
   Max dTM/dt [1/us]:1.81e+03 at an offset of 91.613 usec -> 2021-08-28T07:19:10.561.664.337 (XO: 17.561667191)
   Min dTM/dt [1/us]:-2.14e+03 at an offset of 89.050 usec -> 2021-08-28T07:19:10.561.661.775 (XO: 17.561664628)

 - Block number: 3 starting at 2021-08-28T07:19:36.454.058.725 (XO: 43.454061123 - 43.454367124, t=37.2 degC)  Rank: 00080
   TM values: Max =   188  Min =  -237  Ave =    -4.51  Std =    17.93
   Min TM value:-237 at an offset of 124.338 usec -> 2021-08-28T07:19:36.454.183.062 (XO: 43.454185461)
   Max TM value:188 at an offset of 126.888 usec -> 2021-08-28T07:19:36.454.185.612 (XO: 43.454188011)
   Max dTM/dt [1/us]:1.96e+03 at an offset of 126.700 usec -> 2021-08-28T07:19:36.454.185.425 (XO: 43.454187824)
   Min dTM/dt [1/us]:-2.05e+03 at an offset of 124.125 usec -> 2021-08-28T07:19:36.454.182.850 (XO: 43.454185249)

 - Block number: 4 starting at 2021-08-28T07:19:47.740.670.725 (XO: 54.740674369 - 54.740980370, t=37.2 degC)  Rank: 00082
   TM values: Max =   201  Min =  -248  Ave =    -4.52  Std =    19.62
   Min TM value:-248 at an offset of 40.413 usec -> 2021-08-28T07:19:47.740.711.137 (XO: 54.740714782)
   Max TM value:201 at an offset of 38.375 usec -> 2021-08-28T07:19:47.740.709.100 (XO: 54.740712744)
   Max dTM/dt [1/us]:1.82e+03 at an offset of 38.238 usec -> 2021-08-28T07:19:47.740.708.962 (XO: 54.740712607)
   Min dTM/dt [1/us]:-1.67e+03 at an offset of 27.650 usec -> 2021-08-28T07:19:47.740.698.375 (XO: 54.740702019)

 - Block number: 5 starting at 2021-08-28T07:19:56.867.660.725 (XO: 63.867664488 - 63.867970489, t=37.2 degC)  Rank: 00080
   TM values: Max =   190  Min =  -235  Ave =    -4.43  Std =    19.23
   Min TM value:-235 at an offset of 127.475 usec -> 2021-08-28T07:19:56.867.788.200 (XO: 63.867791964)
   Max TM value:190 at an offset of 130.013 usec -> 2021-08-28T07:19:56.867.790.737 (XO: 63.867794501)
   Max dTM/dt [1/us]:1.91e+03 at an offset of 129.838 usec -> 2021-08-28T07:19:56.867.790.562 (XO: 63.867794326)
   Min dTM/dt [1/us]:-1.98e+03 at an offset of 127.300 usec -> 2021-08-28T07:19:56.867.788.025 (XO: 63.867791789)
