-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Dec 27 09:03:21 2021 UTC
-- File 20210828_071036_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.686104 to 65.941555
-each separately from 4s to 65s and from 173066us to 941555us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.075479 in 1 packets: 18
      12.446730 in 1 packets: 36
      23.963886 in 1 packets: 54
      7.767911 in 1 packets: 72
MIU packet time from 2021-08-28T07:10:43.686.104.000 to 2021-08-28T07:11:44.941.555.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.075479 in 1 packets: 18
      12.446730 in 1 packets: 36
      23.963886 in 1 packets: 54
      7.767911 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.8805612 to 66.8803599
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999439 in 1 packets: 18
      12.9999571 in 1 packets: 36
      23.9999208 in 1 packets: 54
      6.9999769 in 1 packets: 72
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      03 -> 51.1 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 53 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 53 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20210828_071036_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.5666663 to 6.5669553
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.0754230  -- OK for SL --
   MIU time from 2021-08-28T07:10:43.686.104.000 to 2021-08-28T07:10:43.686.393.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.075479

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.6423783 to 23.6426673
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.4466870  -- OK for SL --
   MIU time from 2021-08-28T07:11:00.761.872.000 to 2021-08-28T07:11:00.762.161.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.446730

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.0893543 to 36.0896433
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.9638070  -- OK for SL --
   MIU time from 2021-08-28T07:11:13.208.891.000 to 2021-08-28T07:11:13.209.180.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.963886

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 60.0534503 to 60.0537393
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.7678870  -- OK for SL --
   MIU time from 2021-08-28T07:11:37.173.066.000 to 2021-08-28T07:11:37.173.355.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.767911

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 67.8216263 to 67.8219153
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-28T07:11:44.941.266.000 to 2021-08-28T07:11:44.941.555.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 6.566666300 corrected by 40bit 10MHz counter of 5.880561200 obtained at 1Hz counter of 4. Result: 4.686107433
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-28T07:10:43.685.963.725
   Max 1st sample MIU time among all SL packets: 2021-08-28T07:11:44.941.414.725
   Min 1st sample XO time among all SL packets: 4.685967157
   Max 1st sample XO time among all SL packets: 65.941418232
   TM values from -273 to 221 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-28T07:10:43.685.963.725 (XO: 4.685967157 - 4.686273158, t=37.2 degC)  Rank: 00079
   TM values: Max =   189  Min =  -237  Ave =    -4.52  Std =    19.52
   Min TM value:-237 at an offset of 117.738 usec -> 2021-08-28T07:10:43.686.081.462 (XO: 4.686084895)
   Max TM value:189 at an offset of 120.313 usec -> 2021-08-28T07:10:43.686.084.037 (XO: 4.686087470)
   Max dTM/dt [1/us]:1.64e+03 at an offset of 120.150 usec -> 2021-08-28T07:10:43.686.083.875 (XO: 4.686087308)
   Min dTM/dt [1/us]:-1.98e+03 at an offset of 117.550 usec -> 2021-08-28T07:10:43.686.081.275 (XO: 4.686084708)

 - Block number: 2 starting at 2021-08-28T07:11:00.761.731.725 (XO: 21.761735439 - 21.762041440, t=37.2 degC)  Rank: 00081
   TM values: Max =   221  Min =  -273  Ave =    -4.51  Std =    23.79
   Min TM value:-273 at an offset of 119.788 usec -> 2021-08-28T07:11:00.761.851.512 (XO: 21.761855227)
   Max TM value:221 at an offset of 119.500 usec -> 2021-08-28T07:11:00.761.851.225 (XO: 21.761854939)
   Max dTM/dt [1/us]:2.16e+03 at an offset of 122.100 usec -> 2021-08-28T07:11:00.761.853.825 (XO: 21.761857539)
   Min dTM/dt [1/us]:-2.24e+03 at an offset of 119.588 usec -> 2021-08-28T07:11:00.761.851.312 (XO: 21.761855027)

 - Block number: 3 starting at 2021-08-28T07:11:13.208.750.725 (XO: 34.208752514 - 34.209058515, t=37.2 degC)  Rank: 00083
   TM values: Max =   188  Min =  -234  Ave =    -4.52  Std =    19.38
   Min TM value:-234 at an offset of 104.413 usec -> 2021-08-28T07:11:13.208.855.137 (XO: 34.208856927)
   Max TM value:188 at an offset of 102.313 usec -> 2021-08-28T07:11:13.208.853.037 (XO: 34.208854827)
   Max dTM/dt [1/us]:1.77e+03 at an offset of 102.175 usec -> 2021-08-28T07:11:13.208.852.900 (XO: 34.208854689)
   Min dTM/dt [1/us]:-1.90e+03 at an offset of 104.263 usec -> 2021-08-28T07:11:13.208.854.987 (XO: 34.208856777)

 - Block number: 4 starting at 2021-08-28T07:11:37.172.925.725 (XO: 58.172927596 - 58.173233597, t=37.2 degC)  Rank: 00078
   TM values: Max =   188  Min =  -217  Ave =    -4.57  Std =    17.47
   Min TM value:-217 at an offset of 107.038 usec -> 2021-08-28T07:11:37.173.032.762 (XO: 58.173034634)
   Max TM value:188 at an offset of 107.300 usec -> 2021-08-28T07:11:37.173.033.025 (XO: 58.173034896)
   Max dTM/dt [1/us]:1.99e+03 at an offset of 107.150 usec -> 2021-08-28T07:11:37.173.032.875 (XO: 58.173034746)
   Min dTM/dt [1/us]:-2.02e+03 at an offset of 104.588 usec -> 2021-08-28T07:11:37.173.030.312 (XO: 58.173032184)

 - Block number: 5 starting at 2021-08-28T07:11:44.941.125.725 (XO: 65.941129231 - 65.941435232, t=37.2 degC)  Rank: 00077
   TM values: Max =   201  Min =  -233  Ave =    -4.59  Std =    15.47
   Min TM value:-233 at an offset of 107.950 usec -> 2021-08-28T07:11:44.941.233.675 (XO: 65.941237181)
   Max TM value:201 at an offset of 110.463 usec -> 2021-08-28T07:11:44.941.236.187 (XO: 65.941239694)
   Max dTM/dt [1/us]:1.73e+03 at an offset of 110.300 usec -> 2021-08-28T07:11:44.941.236.025 (XO: 65.941239531)
   Min dTM/dt [1/us]:-1.92e+03 at an offset of 107.788 usec -> 2021-08-28T07:11:44.941.233.512 (XO: 65.941237019)
