-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Dec 25 00:24:23 2021 UTC
-- File 20210825_162512_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.168228 to 43.006022
-each separately from 4s to 43s and from 5733us to 807530us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.638724 in 1 packets: 18
      22.198203 in 1 packets: 36
MIU packet time from 2021-08-25T16:25:19.168.228.000 to 2021-08-25T16:25:58.006.022.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.638724 in 1 packets: 18
      22.198203 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.8992152 to 44.8990859
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999470 in 1 packets: 18
      22.9999237 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210825_162512_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.0674442 to 6.0677332
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.6386710  -- OK for SL --
   MIU time from 2021-08-25T16:25:19.168.228.000 to 2021-08-25T16:25:19.168.517.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.638724

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.7064042 to 22.7066932
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.1981270  -- OK for SL --
   MIU time from 2021-08-25T16:25:35.807.241.000 to 2021-08-25T16:25:35.807.530.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.198203

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 44.9048202 to 44.9051092
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-25T16:25:58.005.733.000 to 2021-08-25T16:25:58.006.022.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 6.067444200 corrected by 40bit 10MHz counter of 5.899215200 obtained at 1Hz counter of 4. Result: 4.168229555
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-25T16:25:19.168.087.725
   Max 1st sample MIU time among all SL packets: 2021-08-25T16:25:58.005.881.725
   Min 1st sample XO time among all SL packets: 4.168089280
   Max 1st sample XO time among all SL packets: 43.005883044
   TM values from -252 to 228 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-25T16:25:19.168.087.725 (XO: 4.168089280 - 4.168395281, t=37.2 degC)  Rank: 00082
   TM values: Max =   228  Min =  -246  Ave =    -4.55  Std =    24.63
   Min TM value:-246 at an offset of 127.825 usec -> 2021-08-25T16:25:19.168.215.550 (XO: 4.168217105)
   Max TM value:228 at an offset of 130.463 usec -> 2021-08-25T16:25:19.168.218.187 (XO: 4.168219743)
   Max dTM/dt [1/us]:1.63e+03 at an offset of 130.263 usec -> 2021-08-25T16:25:19.168.217.987 (XO: 4.168219543)
   Min dTM/dt [1/us]:-2.12e+03 at an offset of 127.663 usec -> 2021-08-25T16:25:19.168.215.387 (XO: 4.168216943)

 - Block number: 2 starting at 2021-08-25T16:25:35.807.100.725 (XO: 20.807104388 - 20.807410389, t=37.2 degC)  Rank: 00078
   TM values: Max =   216  Min =  -217  Ave =    -4.56  Std =    18.52
   Min TM value:-217 at an offset of 115.400 usec -> 2021-08-25T16:25:35.807.216.125 (XO: 20.807219789)
   Max TM value:216 at an offset of 118.163 usec -> 2021-08-25T16:25:35.807.218.887 (XO: 20.807222551)
   Max dTM/dt [1/us]:1.76e+03 at an offset of 118.000 usec -> 2021-08-25T16:25:35.807.218.725 (XO: 20.807222389)
   Min dTM/dt [1/us]:-1.59e+03 at an offset of 115.213 usec -> 2021-08-25T16:25:35.807.215.937 (XO: 20.807219601)

 - Block number: 3 starting at 2021-08-25T16:25:58.005.592.725 (XO: 43.005594043 - 43.005900044, t=37.2 degC)  Rank: 00076
   TM values: Max =   210  Min =  -252  Ave =    -4.50  Std =    18.21
   Min TM value:-252 at an offset of 116.075 usec -> 2021-08-25T16:25:58.005.708.800 (XO: 43.005710119)
   Max TM value:210 at an offset of 118.650 usec -> 2021-08-25T16:25:58.005.711.375 (XO: 43.005712694)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 118.463 usec -> 2021-08-25T16:25:58.005.711.187 (XO: 43.005712506)
   Min dTM/dt [1/us]:-2.00e+03 at an offset of 115.888 usec -> 2021-08-25T16:25:58.005.708.612 (XO: 43.005709931)
