-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Dec 24 23:17:48 2021 UTC
-- File 20210825_154012_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.304283 to 19.684189
-each separately from 14s to 19s and from 304283us to 684189us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.379328 in 1 packets: 18
MIU packet time from 2021-08-25T15:40:29.304.283.000 to 2021-08-25T15:40:34.684.189.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.379328 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.8977661 to 20.8977495
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.9999834 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210825_154012_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.2020500 to 16.2023390
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.3793110  -- OK for SL --
   MIU time from 2021-08-25T15:40:29.304.283.000 to 2021-08-25T15:40:29.304.572.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.379328

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.5816500 to 21.5819390
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-25T15:40:34.683.900.000 to 2021-08-25T15:40:34.684.189.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 16.202050000 corrected by 40bit 10MHz counter of 15.897766100 obtained at 1Hz counter of 14. Result: 14.304284935
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-08-25T15:40:29.304.142.725
   Max 1st sample MIU time among all SL packets: 2021-08-25T15:40:34.684.048.725
   Min 1st sample XO time among all SL packets: 14.304144659
   Max 1st sample XO time among all SL packets: 19.684051551
   TM values from -325 to 258 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-25T15:40:29.304.142.725 (XO: 14.304144659 - 14.304450660, t=37.2 degC)  Rank: 00083
   TM values: Max =   245  Min =  -282  Ave =    -4.54  Std =    23.69
   Min TM value:-282 at an offset of 110.388 usec -> 2021-08-25T15:40:29.304.253.112 (XO: 14.304255047)
   Max TM value:245 at an offset of 110.125 usec -> 2021-08-25T15:40:29.304.252.850 (XO: 14.304254784)
   Max dTM/dt [1/us]:2.39e+03 at an offset of 112.825 usec -> 2021-08-25T15:40:29.304.255.550 (XO: 14.304257484)
   Min dTM/dt [1/us]:-2.74e+03 at an offset of 110.275 usec -> 2021-08-25T15:40:29.304.253.000 (XO: 14.304254934)

 - Block number: 2 starting at 2021-08-25T15:40:34.683.759.725 (XO: 19.683762550 - 19.684068551, t=37.2 degC)  Rank: 00083
   TM values: Max =   258  Min =  -325  Ave =    -4.52  Std =    32.17
   Min TM value:-325 at an offset of 122.113 usec -> 2021-08-25T15:40:34.683.881.837 (XO: 19.683884663)
   Max TM value:258 at an offset of 121.863 usec -> 2021-08-25T15:40:34.683.881.587 (XO: 19.683884413)
   Max dTM/dt [1/us]:2.25e+03 at an offset of 119.900 usec -> 2021-08-25T15:40:34.683.879.625 (XO: 19.683882450)
   Min dTM/dt [1/us]:-2.98e+03 at an offset of 121.988 usec -> 2021-08-25T15:40:34.683.881.712 (XO: 19.683884538)
