-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Dec 24 18:12:10 2021 UTC
-- File 20210825_151238_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.455765 to 24.461831
-each separately from 14s to 24s and from 455765us to 461831us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.005488 in 1 packets: 18
MIU packet time from 2021-08-25T15:12:55.455.765.000 to 2021-08-25T15:13:05.461.831.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.005488 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.9368699 to 25.9368368
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999669 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210825_151238_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.3926359 to 16.3929249
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.0054550  -- OK for SL --
   MIU time from 2021-08-25T15:12:55.455.765.000 to 2021-08-25T15:12:55.456.054.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.005488

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.3983799 to 26.3986689
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-25T15:13:05.461.542.000 to 2021-08-25T15:13:05.461.831.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 16.392635900 corrected by 40bit 10MHz counter of 15.936869900 obtained at 1Hz counter of 14. Result: 14.455767504
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-25T15:12:55.455.624.725
   Max 1st sample MIU time among all SL packets: 2021-08-25T15:13:05.461.690.725
   Min 1st sample XO time among all SL packets: 14.455627229
   Max 1st sample XO time among all SL packets: 24.461693349
   TM values from -229 to 236 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-25T15:12:55.455.624.725 (XO: 14.455627229 - 14.455933230, t=37.2 degC)  Rank: 00077
   TM values: Max =   236  Min =  -229  Ave =    -4.49  Std =    20.34
   Min TM value:-229 at an offset of 110.113 usec -> 2021-08-25T15:12:55.455.734.837 (XO: 14.455737341)
   Max TM value:236 at an offset of 112.825 usec -> 2021-08-25T15:12:55.455.737.550 (XO: 14.455740054)
   Max dTM/dt [1/us]:1.85e+03 at an offset of 112.713 usec -> 2021-08-25T15:12:55.455.737.437 (XO: 14.455739941)
   Min dTM/dt [1/us]:-1.82e+03 at an offset of 109.988 usec -> 2021-08-25T15:12:55.455.734.712 (XO: 14.455737216)

 - Block number: 2 starting at 2021-08-25T15:13:05.461.401.725 (XO: 24.461404348 - 24.461710349, t=37.2 degC)  Rank: 00085
   TM values: Max =   220  Min =  -219  Ave =    -4.50  Std =    20.81
   Min TM value:-219 at an offset of 119.050 usec -> 2021-08-25T15:13:05.461.520.775 (XO: 24.461523398)
   Max TM value:220 at an offset of 121.713 usec -> 2021-08-25T15:13:05.461.523.437 (XO: 24.461526061)
   Max dTM/dt [1/us]:2.01e+03 at an offset of 121.588 usec -> 2021-08-25T15:13:05.461.523.312 (XO: 24.461525936)
   Min dTM/dt [1/us]:-1.86e+03 at an offset of 118.825 usec -> 2021-08-25T15:13:05.461.520.550 (XO: 24.461523173)
