-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Dec 24 15:58:04 2021 UTC
-- File 20210825_115633_1227.dat, 18  packets found.
APID:
      1227 (SL) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.293753 to 8.294042
-each separately from 8s to 8s and from 293753us to 294042us.
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU packet time from 2021-08-25T11:56:44.293.753.000 to 2021-08-25T11:56:44.294.042.000
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.8813291 to 9.8813291
   Differences in neigboring packets: 
      0.0000000 in 17 packets (all packets)
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 18 packets (all packets)
Number of generated alerts: 
      0 in 18 packets (all packets)
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20210825_115633_1227.dat, 1 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.1750831 to 10.1753721
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-25T11:56:44.293.753.000 to 2021-08-25T11:56:44.294.042.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 10.175083100 corrected by 40bit 10MHz counter of 9.881329100 obtained at 1Hz counter of 8. Result: 8.293754940
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996800 - 0.999996800
   Min 1st sample MIU time among all SL packets: 2021-08-25T11:56:44.293.612.725
   Max 1st sample MIU time among all SL packets: 2021-08-25T11:56:44.293.901.725
   Min 1st sample XO time among all SL packets: 8.293614665
   Max 1st sample XO time among all SL packets: 8.293903665
   TM values from -232 to 232 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-25T11:56:44.293.612.725 (XO: 8.293614665 - 8.293920666, t=35.6 degC)  Rank: 00080
   TM values: Max =   232  Min =  -232  Ave =    -4.57  Std =    22.24
   Min TM value:-232 at an offset of 124.963 usec -> 2021-08-25T11:56:44.293.737.687 (XO: 8.293739627)
   Max TM value:232 at an offset of 127.613 usec -> 2021-08-25T11:56:44.293.740.337 (XO: 8.293742277)
   Max dTM/dt [1/us]:1.79e+03 at an offset of 127.438 usec -> 2021-08-25T11:56:44.293.740.162 (XO: 8.293742102)
   Min dTM/dt [1/us]:-1.85e+03 at an offset of 124.775 usec -> 2021-08-25T11:56:44.293.737.500 (XO: 8.293739440)
