-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Dec 23 23:57:09 2021 UTC
-- File 20210824_101416_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.185295 to 16.275841
-each separately from 6s to 16s and from 185295us to 275841us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.089968 in 1 packets: 18
MIU packet time from 2021-08-24T10:14:25.185.295.000 to 2021-08-24T10:14:35.275.841.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.089968 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.8726979 to 17.8726647
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999668 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210824_101416_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.0579942 to 8.0582832
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.0899350  -- OK for SL --
   MIU time from 2021-08-24T10:14:25.185.295.000 to 2021-08-24T10:14:25.185.584.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.089968

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.1482182 to 18.1485072
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-24T10:14:35.275.552.000 to 2021-08-24T10:14:35.275.841.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.057994200 corrected by 40bit 10MHz counter of 7.872697900 obtained at 1Hz counter of 6. Result: 6.185296930
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-24T10:14:25.185.154.725
   Max 1st sample MIU time among all SL packets: 2021-08-24T10:14:35.275.700.725
   Min 1st sample XO time among all SL packets: 6.185156655
   Max 1st sample XO time among all SL packets: 16.275703135
   TM values from -258 to 192 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-24T10:14:25.185.154.725 (XO: 6.185156655 - 6.185462656, t=37.2 degC)  Rank: 00077
   TM values: Max =   191  Min =  -258  Ave =    -4.45  Std =    18.48
   Min TM value:-258 at an offset of 127.638 usec -> 2021-08-24T10:14:25.185.282.362 (XO: 6.185284292)
   Max TM value:191 at an offset of 127.325 usec -> 2021-08-24T10:14:25.185.282.050 (XO: 6.185283980)
   Max dTM/dt [1/us]:1.69e+03 at an offset of 130.000 usec -> 2021-08-24T10:14:25.185.284.725 (XO: 6.185286655)
   Min dTM/dt [1/us]:-1.82e+03 at an offset of 127.413 usec -> 2021-08-24T10:14:25.185.282.137 (XO: 6.185284067)

 - Block number: 2 starting at 2021-08-24T10:14:35.275.411.725 (XO: 16.275414134 - 16.275720135, t=37.2 degC)  Rank: 00083
   TM values: Max =   192  Min =  -237  Ave =    -4.52  Std =    22.60
   Min TM value:-237 at an offset of 124.200 usec -> 2021-08-24T10:14:35.275.535.925 (XO: 16.275538334)
   Max TM value:192 at an offset of 126.888 usec -> 2021-08-24T10:14:35.275.538.612 (XO: 16.275541022)
   Max dTM/dt [1/us]:1.76e+03 at an offset of 126.750 usec -> 2021-08-24T10:14:35.275.538.475 (XO: 16.275540884)
   Min dTM/dt [1/us]:-1.61e+03 at an offset of 124.050 usec -> 2021-08-24T10:14:35.275.535.775 (XO: 16.275538184)
