-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Dec 23 23:11:30 2021 UTC
-- File 20210824_093419_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.533627 to 63.172087
-each separately from 6s to 63s and from 52386us to 972952us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.309057 in 1 packets: 18
      8.129401 in 1 packets: 36
      12.079434 in 1 packets: 54
      16.119123 in 1 packets: 72
MIU packet time from 2021-08-24T09:34:28.533.627.000 to 2021-08-24T09:35:25.172.087.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.309057 in 1 packets: 18
      8.129401 in 1 packets: 36
      12.079434 in 1 packets: 54
      16.119123 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.9275908 to 64.9274034
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.9999342 in 1 packets: 18
      7.9999737 in 1 packets: 36
      12.9999573 in 1 packets: 54
      15.9999474 in 1 packets: 72
Temperature FPGA: 
      05 -> 48.5 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 71 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 71 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20210824_093419_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.4612188 to 8.4615078
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.3089910  -- OK for SL --
   MIU time from 2021-08-24T09:34:28.533.627.000 to 2021-08-24T09:34:28.533.916.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.309057

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.7704988 to 28.7707878
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.1293750  -- OK for SL --
   MIU time from 2021-08-24T09:34:48.842.973.000 to 2021-08-24T09:34:48.843.262.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.129401

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      73 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.9001628 to 36.9004518
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.0793910  -- OK for SL --
   MIU time from 2021-08-24T09:34:56.972.663.000 to 2021-08-24T09:34:56.972.952.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.079434

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 48.9798428 to 48.9801318
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.1190710  -- OK for SL --
   MIU time from 2021-08-24T09:35:09.052.386.000 to 2021-08-24T09:35:09.052.675.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.119123

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 65.0992028 to 65.0994918
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-24T09:35:25.171.798.000 to 2021-08-24T09:35:25.172.087.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.461218800 corrected by 40bit 10MHz counter of 7.927590800 obtained at 1Hz counter of 6. Result: 6.533629761
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-24T09:34:28.533.486.725
   Max 1st sample MIU time among all SL packets: 2021-08-24T09:35:25.171.946.725
   Min 1st sample XO time among all SL packets: 6.533489486
   Max 1st sample XO time among all SL packets: 63.171948692
   TM values from -243 to 203 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-24T09:34:28.533.486.725 (XO: 6.533489486 - 6.533795487, t=37.2 degC)  Rank: 00079
   TM values: Max =   203  Min =  -240  Ave =    -4.41  Std =    18.99
   Min TM value:-240 at an offset of 114.100 usec -> 2021-08-24T09:34:28.533.600.825 (XO: 6.533603586)
   Max TM value:203 at an offset of 113.800 usec -> 2021-08-24T09:34:28.533.600.525 (XO: 6.533603286)
   Max dTM/dt [1/us]:2.05e+03 at an offset of 116.450 usec -> 2021-08-24T09:34:28.533.603.175 (XO: 6.533605936)
   Min dTM/dt [1/us]:-2.16e+03 at an offset of 113.900 usec -> 2021-08-24T09:34:28.533.600.625 (XO: 6.533603386)

 - Block number: 2 starting at 2021-08-24T09:34:48.842.832.725 (XO: 26.842836306 - 26.843142307, t=37.2 degC)  Rank: 00081
   TM values: Max =   170  Min =  -225  Ave =    -4.50  Std =    22.14
   Min TM value:-225 at an offset of 123.425 usec -> 2021-08-24T09:34:48.842.956.150 (XO: 26.842959732)
   Max TM value:170 at an offset of 126.050 usec -> 2021-08-24T09:34:48.842.958.775 (XO: 26.842962357)
   Max dTM/dt [1/us]:1.44e+03 at an offset of 125.938 usec -> 2021-08-24T09:34:48.842.958.662 (XO: 26.842962244)
   Min dTM/dt [1/us]:-1.64e+03 at an offset of 110.325 usec -> 2021-08-24T09:34:48.842.943.050 (XO: 26.842946632)

 - Block number: 3 starting at 2021-08-24T09:34:56.972.522.725 (XO: 34.972527034 - 34.972833035, t=37.2 degC)  Rank: 00073
   TM values: Max =   198  Min =  -243  Ave =    -4.43  Std =    28.45
   Min TM value:-243 at an offset of 199.851 usec -> 2021-08-24T09:34:56.972.722.575 (XO: 34.972726885)
   Max TM value:198 at an offset of 197.588 usec -> 2021-08-24T09:34:56.972.720.313 (XO: 34.972724622)
   Max dTM/dt [1/us]:1.62e+03 at an offset of 197.426 usec -> 2021-08-24T09:34:56.972.720.150 (XO: 34.972724460)
   Min dTM/dt [1/us]:-1.86e+03 at an offset of 227.501 usec -> 2021-08-24T09:34:56.972.750.225 (XO: 34.972754535)

 - Block number: 4 starting at 2021-08-24T09:35:09.052.245.725 (XO: 47.052246697 - 47.052552698, t=37.2 degC)  Rank: 00076
   TM values: Max =   178  Min =  -238  Ave =    -4.39  Std =    24.11
   Min TM value:-238 at an offset of 124.250 usec -> 2021-08-24T09:35:09.052.369.975 (XO: 47.052370948)
   Max TM value:178 at an offset of 97.063 usec -> 2021-08-24T09:35:09.052.342.787 (XO: 47.052343760)
   Max dTM/dt [1/us]:1.69e+03 at an offset of 99.750 usec -> 2021-08-24T09:35:09.052.345.475 (XO: 47.052346448)
   Min dTM/dt [1/us]:-1.68e+03 at an offset of 112.138 usec -> 2021-08-24T09:35:09.052.357.862 (XO: 47.052358835)

 - Block number: 5 starting at 2021-08-24T09:35:25.171.657.725 (XO: 63.171659691 - 63.171965692, t=37.2 degC)  Rank: 00080
   TM values: Max =   179  Min =  -223  Ave =    -4.46  Std =    18.01
   Min TM value:-223 at an offset of 111.625 usec -> 2021-08-24T09:35:25.171.769.350 (XO: 63.171771317)
   Max TM value:179 at an offset of 114.250 usec -> 2021-08-24T09:35:25.171.771.975 (XO: 63.171773942)
   Max dTM/dt [1/us]:1.52e+03 at an offset of 114.050 usec -> 2021-08-24T09:35:25.171.771.775 (XO: 63.171773742)
   Min dTM/dt [1/us]:-1.81e+03 at an offset of 111.438 usec -> 2021-08-24T09:35:25.171.769.162 (XO: 63.171771129)
