-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Dec 23 13:54:24 2021 UTC
-- File 20210823_163915_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.243047 to 44.646237
-each separately from 9s to 44s and from 43483us to 646237us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.800147 in 1 packets: 18
      29.602176 in 1 packets: 36
MIU packet time from 2021-08-23T16:39:27.243.047.000 to 2021-08-23T16:40:02.646.237.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.800147 in 1 packets: 18
      29.602176 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.8394554 to 45.8393389
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.9999800 in 1 packets: 18
      28.9999035 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      03 -> 51.1 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      3 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      3 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210823_163915_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.0825033 to 11.0827923
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.8001270  -- OK for SL --
   MIU time from 2021-08-23T16:39:27.243.047.000 to 2021-08-23T16:39:27.243.336.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.800147

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      72 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.8829193 to 16.8832083
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 29.6020790  -- OK for SL --
   MIU time from 2021-08-23T16:39:33.043.483.000 to 2021-08-23T16:39:33.043.772.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 29.602176

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.4852873 to 46.4855763
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-23T16:40:02.645.948.000 to 2021-08-23T16:40:02.646.237.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.082503300 corrected by 40bit 10MHz counter of 10.839455400 obtained at 1Hz counter of 9. Result: 9.243048726
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-23T16:39:27.242.906.725
   Max 1st sample MIU time among all SL packets: 2021-08-23T16:40:02.646.096.725
   Min 1st sample XO time among all SL packets: 9.242908451
   Max 1st sample XO time among all SL packets: 44.646099257
   TM values from -259 to 213 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-23T16:39:27.242.906.725 (XO: 9.242908451 - 9.243214452, t=37.2 degC)  Rank: 00087
   TM values: Max =   213  Min =  -259  Ave =    -4.49  Std =    21.53
   Min TM value:-259 at an offset of 119.638 usec -> 2021-08-23T16:39:27.243.026.362 (XO: 9.243028089)
   Max TM value:213 at an offset of 119.300 usec -> 2021-08-23T16:39:27.243.026.025 (XO: 9.243027751)
   Max dTM/dt [1/us]:2.20e+03 at an offset of 121.988 usec -> 2021-08-23T16:39:27.243.028.712 (XO: 9.243030439)
   Min dTM/dt [1/us]:-2.47e+03 at an offset of 119.413 usec -> 2021-08-23T16:39:27.243.026.137 (XO: 9.243027864)

 - Block number: 2 starting at 2021-08-23T16:39:33.043.342.725 (XO: 15.043343768 - 15.043649769, t=37.2 degC)  Rank: 00072
   TM values: Max =   197  Min =  -257  Ave =    -4.42  Std =    26.68
   Min TM value:-257 at an offset of 193.426 usec -> 2021-08-23T16:39:33.043.536.150 (XO: 15.043537194)
   Max TM value:197 at an offset of 122.075 usec -> 2021-08-23T16:39:33.043.464.800 (XO: 15.043465843)
   Max dTM/dt [1/us]:1.70e+03 at an offset of 121.888 usec -> 2021-08-23T16:39:33.043.464.612 (XO: 15.043465656)
   Min dTM/dt [1/us]:-2.15e+03 at an offset of 193.326 usec -> 2021-08-23T16:39:33.043.536.050 (XO: 15.043537094)

 - Block number: 3 starting at 2021-08-23T16:40:02.645.807.725 (XO: 44.645810256 - 44.646116257, t=37.2 degC)  Rank: 00082
   TM values: Max =   208  Min =  -250  Ave =    -4.42  Std =    24.21
   Min TM value:-250 at an offset of 120.600 usec -> 2021-08-23T16:40:02.645.928.325 (XO: 44.645930857)
   Max TM value:208 at an offset of 123.100 usec -> 2021-08-23T16:40:02.645.930.825 (XO: 44.645933357)
   Max dTM/dt [1/us]:2.43e+03 at an offset of 122.938 usec -> 2021-08-23T16:40:02.645.930.662 (XO: 44.645933194)
   Min dTM/dt [1/us]:-2.32e+03 at an offset of 120.425 usec -> 2021-08-23T16:40:02.645.928.150 (XO: 44.645930682)
