-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Dec 23 11:43:37 2021 UTC
-- File 20210823_150050_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.227636 to 21.318156
-each separately from 9s to 21s and from 227636us to 318156us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.089942 in 1 packets: 18
MIU packet time from 2021-08-23T15:01:02.227.636.000 to 2021-08-23T15:01:14.318.156.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.089942 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.8903996 to 22.8903601
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999605 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210823_150050_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.1180365 to 11.1183255
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.0899030  -- OK for SL --
   MIU time from 2021-08-23T15:01:02.227.636.000 to 2021-08-23T15:01:02.227.925.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.089942

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.2082285 to 23.2085175
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-23T15:01:14.317.867.000 to 2021-08-23T15:01:14.318.156.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.118036500 corrected by 40bit 10MHz counter of 10.890399600 obtained at 1Hz counter of 9. Result: 9.227637651
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-23T15:01:02.227.495.725
   Max 1st sample MIU time among all SL packets: 2021-08-23T15:01:14.318.015.725
   Min 1st sample XO time among all SL packets: 9.227497376
   Max 1st sample XO time among all SL packets: 21.318018206
   TM values from -252 to 201 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-23T15:01:02.227.495.725 (XO: 9.227497376 - 9.227803377, t=37.2 degC)  Rank: 00088
   TM values: Max =   196  Min =  -252  Ave =    -4.45  Std =    31.27
   Min TM value:-252 at an offset of 218.938 usec -> 2021-08-23T15:01:02.227.714.663 (XO: 9.227716314)
   Max TM value:196 at an offset of 120.788 usec -> 2021-08-23T15:01:02.227.616.512 (XO: 9.227618164)
   Max dTM/dt [1/us]:1.71e+03 at an offset of 221.413 usec -> 2021-08-23T15:01:02.227.717.138 (XO: 9.227718789)
   Min dTM/dt [1/us]:-1.95e+03 at an offset of 218.826 usec -> 2021-08-23T15:01:02.227.714.550 (XO: 9.227716201)

 - Block number: 2 starting at 2021-08-23T15:01:14.317.726.725 (XO: 21.317729205 - 21.318035206, t=37.2 degC)  Rank: 00079
   TM values: Max =   201  Min =  -248  Ave =    -4.55  Std =    22.35
   Min TM value:-248 at an offset of 121.838 usec -> 2021-08-23T15:01:14.317.848.562 (XO: 21.317851043)
   Max TM value:201 at an offset of 182.426 usec -> 2021-08-23T15:01:14.317.909.150 (XO: 21.317911631)
   Max dTM/dt [1/us]:1.81e+03 at an offset of 182.313 usec -> 2021-08-23T15:01:14.317.909.038 (XO: 21.317911518)
   Min dTM/dt [1/us]:-1.96e+03 at an offset of 121.700 usec -> 2021-08-23T15:01:14.317.848.425 (XO: 21.317850906)
